The Project Gutenberg eBook of Refraction and muscular imbalance, as simplified through the use of the ski-optometer

This eBook is for the use of anyone anywhere in the United States and most other parts of the world at no cost and with almost no restrictions whatsoever. You may copy it, give it away or re-use it under the terms of the Project Gutenberg License included with this eBook or online at www.gutenberg.org. If you are not located in the United States, you will have to check the laws of the country where you are located before using this eBook.

Title: Refraction and muscular imbalance, as simplified through the use of the ski-optometer

Author: Daniel Woolf

Release date: August 29, 2023 [eBook #71517]

Language: English

Original publication: New York: Theodore S. Holbrook, 1921

Other information and formats: www.gutenberg.org/ebooks/71517

Credits: deaurider and the Online Distributed Proofreading Team at https://www.pgdp.net (This file was produced from images generously made available by The Internet Archive)

*** START OF THE PROJECT GUTENBERG EBOOK REFRACTION AND MUSCULAR IMBALANCE, AS SIMPLIFIED THROUGH THE USE OF THE SKI-OPTOMETER ***

SKI-OPTOMETER MASTER MODEL 215

Embodying in a Single Instrument, in Convenient Form,

Cylindrical and Spherical Lenses, in Combination

with Appliances for Testing and Correcting

Muscular Imbalance.

Refraction and

Muscular Imbalance

As Simplified Through the Use

of the Ski-optometer

By

DANIEL WOOLF

WOOLF INSTRUMENT CORPORATION

NEW YORK: 516 FIFTH AVENUE

Copyright 1921

By WOOLF INSTRUMENT CORPORATION

Published by

THEODORE S. HOLBROOK

NEW YORK

CONTENTS

Page

CHAPTER I

SKI-OPTOMETER CONSTRUCTION

1

Convex Spherical Lenses

2

Operates and Indicates Automatically

6

Concave Spherical Lenses

7

CHAPTER II

CYLINDRICAL LENSES

10

Obtaining Correct Focus

11

Why Concave Cylinders Are Used Exclusively

14

Transposition of Lenses

14

CHAPTER III

HOW THE SKI-OPTOMETER ASSISTS IN REFRACTION

17

The Use of the Ski-optometer in Skioscopy

17

A Simplified Skioscopic Method

20

Employing Spheres and Cylinders in Skioscopy

22

Use of the Ski-optometer in Subjective Testing

23

A Simplified Subjective Method

24

Procedure for Using Minus Cylinders Exclusively

26

Constant Attention Not Required

29

CHAPTER IV

IMPORTANT POINTS IN CONNECTION WITH THE

USE OF THE SKI-OPTOMETER

30

Elimination of Trial-Frame Discomfort

30

Rigidity of Construction

31

How to Place the Ski-optometer in Position

32

Cleaning the Lenses

33

Accuracy Assured in Every Test

34

Built to Last a Lifetime

35

CHAPTER V

CONDENSED PROCEDURE FOR MAKING SPHERE AND

CYLINDER TEST WITH THE SKI-OPTOMETER

37

Subjective Distance Test

37

Subjective Reading Test

40

CHAPTER VI

MUSCULAR IMBALANCE

41

The Action of Prisms

42

The Phorometer

43

The Maddox Rod

44

Procedure for Making the Muscle Test

45

Binocular and Monocular Test

47

CHAPTER VII

THE BINOCULAR MUSCLE TEST

48

Made with the Maddox Rod and Phorometer

48

Esophoria and Exophoria

50

Making Muscle Test Before and After Optical Correction

52

When to Consider Correction of Muscular Imbalance

53

Four Methods for Correction of Muscular Imbalance

54

The Rotary Prism

54

Use of the Rotary Prism in Binocular Muscle Tests

56

CHAPTER VIII

THE MONOCULAR DUCTION MUSCLE TEST

58

Made with Both Rotary Prisms

58

Locating the Faulty Muscle

58

Adduction

59

Abduction

61

Superduction

62

Subduction

63

Procedure for Monocular Muscle Testing

64

Diagnosing a Specific Muscle Case

65

CHAPTER IX

FIRST METHOD OF TREATMENT—OPTICAL CORRECTION

70

Esophoria

70

Treatment for Correcting Esophoria in Children

72

How Optical Correction Tends to Decrease 6°

Esophoria in a Child

74

CHAPTER X

SECOND METHOD OF TREATMENT—MUSCULAR EXERCISE

75

Made with Two Rotary Prisms and Red Maddox Rod

75

Exophoria

75

An Assumed Case

78

Effect of Muscular Exercise

80

Home Treatment for Muscular Exercise—

Square Prism Set Used in Conjunction with

the Ski-optometer

82

CHAPTER XI

THIRD METHOD OF TREATMENT—PRISM LENSES

84

When and How Employed

84

Prism Reduction Method

85

CHAPTER XII

A CONDENSATION OF PREVIOUS CHAPTERS ON THE PROCEDURE

FOR MUSCLE TESTING WITH THE SKI-OPTOMETER

87

Four Methods of Treating an Imbalance Case when

the Preceding One Fails

90

Prisms

92

Cyclophoria

92

CHAPTER XIII

CYCLOPHORIA

93

Made with Maddox Rods and Rotary Prisms

93

CHAPTER XIV

CYCLODUCTION TEST

99

Made with the Combined Use of the Two Maddox Rods

99

Treatment for Cyclophoria

102

CHAPTER XV

MOVEMENTS OF THE EYEBALLS AND THEIR ANOMALIES

105

Monocular Fixation

105

Binocular Fixation

106

Orthophoria

107

Heterophoria

107

Squint

108

Varieties of Heterophoria and Squint

109

CHAPTER XVI

LAW OF PROJECTION

114

Suppression of Image

115

Monocular Diplopia

115

Table of Diplopia

116

Movement of Each Eye Singly

117

Subsidiary Actions

118

Field of Action of Muscles

120

Direction of the Gaze

120

Primary Position—Field of Fixation

121

Binocular Movements

121

Parallel Movements

122

Lateral Rotators

123

Eye Associates

124

Movements of Convergence

125

Movements of Divergence

125

Vertical Divergence

126

Orthophoria

126

Heterophoria

126

Subdivisions

126

CHAPTER XVII

SYMPTOMS OF HETEROPHORIA

128

Treatment

130

Destrophoria and Laevophoria

132

The demands of the day for maximum efficiency in the refracting world are largely accountable for the inception, continuous improvement and ultimate development of the master model Ski-optometer.

The present volume, dealing with the instrument’s distinctive operative features, has been prepared not only for Ski-optometer users, but also for those interested in the simplification of refraction and muscular imbalance.

The author is indebted for invaluable counsel, to

Louis J. Ameno, M.D., New York.

E. LeRoy Ryer, O.D., New York.

Jos. D. Heitger, M.D., Louisville, Ky.

W. B. Needles, N.D., Kansas City, Mo.

INTRODUCTORY

WHILE in a measure the conventional trial-case still serves its purpose, so much of the refractionist’s time is consumed through the mechanical process of individually transferring the trial-case spheres and cylinder lenses, that far too little thought is given to muscular imbalance, notwithstanding its importance in all refraction cases.

Dr. Samuel Theibold, of Johns Hopkins University, in a recent address before the American Medical Association, stated that the average refractionist was inclined to devote an excess of time to general refraction, completely overlooking the important test and correction of muscular imbalance. If the latter is to be at all considered, general refraction must be simplified—without impairing its accuracy—a result that is greatly facilitated through the use of the Ski-optometer.

One must admit that tediously selecting the required trial-case lens—whether sphere, cylinder or prism—watching the stamped number on the handle—continual wiping and inserting each individual lens in a trial-frame is a time-consuming practise. This is readily overcome, however, through the employment of the Ski-optometer.

In a word, the Ski-optometer is practically an automatic trial-case, bearing the same relation to the refracting room as the accepted labor and time-saving devices of the day bear to the commercial world.

The present volume has accordingly been published, not alone in the interest of those possessing a Ski-optometer, but also for those interested in attaining the highest point of efficiency in the work of refraction and muscular imbalance.

Ski-optometer Lens Battery (almost actual size)

showing how sphere and cylinder lenses are procured.

After obtaining FINAL results, your prescription is automatically registered,

ALL READY for you to transcribe.

Fig. 1—The three time-saving moves necessary in the operation of the Ski-optometer.

CHAPTER I

SKI-OPTOMETER CONSTRUCTION

A FAR better understanding of the instrument will be secured if the refractionist possessing a Ski-optometer will place it before him, working out each operation and experiment step by step in its proper routine.

The three moves as outlined in Fig. 1 should first be thoughtfully studied and the method of obtaining the spheres and cylinders carefully observed.

Fig. 2—To Obtain Plano.

1—

Set spherical indicator at “000” as illustrated above.

2—

Set cylinder indicator to “0”.

3—

Set pointer of supplementary disk at “open”.

The instrument should then be set at zero or “plano,” a position indicated by the appearance of the three “0 0 0” at the spherical register, in conjunction with one “0” or zero, for the cylinder at its register, marked “CC Cyl.”

After this move, the supplementary disk’s pointer should be set at “open” (Fig. 2).

Fig. 3—To obtain sphericals, turn this Single Reel as shown by dotted finger. This assures an automatic and simultaneous registration at sphere indicator of focus of lens appearing at sight opening.

CONVEX SPHERICAL LENSES

A careful study will show that the Ski-optometer’s spherical lenses are obtained by merely turning the smaller reel (Fig. 3). The first outward turn of this reel, toward the temporal side of the instrument, draws into position in regular order the spherical lenses +.25, +.50, +.75, and +1.D., as shown in Fig. 3a.

3-A—Outer spherical reel containing Cx. sphericals

from .025 to 1.00D and a blank.

3-B—Inner spherical disk containing Cx. sphericals,

automatically turns within 3-A.

3-C—Supplementary spherical disk.

By means of a concealed tooth gear, an inner disk is automatically picked up, placing its first lens +1.25D in position (Fig. 3b). This +1.25D spherical lens remains stationary while the outer disk again revolves, adding to it the original +.25, +.50, +.75 and +1.D., the latter totalling +2.25D. At this point, the instrument again automatically picks up its inner disk, thereby placing its second lens, +2.50D, in position.

Fig. 4—With the reappearance of “00” at sphere indicator, a rapid increase or decrease of +1.25 is accurately and speedily attained.

Instead of using intermediate strengths in making an examination, it is frequently desirable to make such extended changes as 1.25D to 2.50D. With the Ski-optometer, the refractionist will note that two white zeros appeared at the spherical register in connection with +1.25, and again with +2.50. A rapid outward turn of the spherical reel toward the temporal side to the point of the reappearance of the two zeros will show +3.75D; or, if increased power is still desired, a rapid turn will draw +5.D. into position (Fig. 4).

Turning the reel inward toward the nasal side will likewise decrease its convex power. In brief, each one of these lenses, showing their foci in conjunction with the two white zeros, are signals indicating the rapid increase or decrease of one and one-quarter diopter. After continuing to +6D., the next turn automatically shows zero (or “plano”), the original starting point, which is again indicated by the three white zeros.

Through the turn of the single reel—an exclusive Ski-optometer feature—all convex spherical lenses have now been attained in quarters up to +6.D, practically covering ninety percent of all refraction cases.

Fig. 5—With supplementary disk pointer set at +6 Sph., this places an additional +6.D spherical lens at sight opening, extending instrument’s total convex spherical power to +12.D.

Should still greater power be desired, the small pointer at the outer edge of the instrument should be set at +6 sphere (Fig. 5). This controls a supplementary disk (Fig. 3c) which places an additional +6D. lens before the original range of lenses previously referred to, thus increasing the maximum power to +12D. If still greater strength is required, any additional trial-case lens may be added, a cell being provided for that purpose on the forward plate of the instrument.

OPERATES AND INDICATES AUTOMATICALLY

As previously explained, in using the Ski-optometer, it is only necessary to remember that each outward turn of the single reel toward the temporal side of the patient increases the plus power, while the reverse turn toward the patient’s nose decreases it. In fact, no attention need ever be given the register until the required sum-total is secured, it only being necessary to turn the single reel in order to be assured of the unvarying and accurate operation of the instrument.

For convenience, the contour or upper edge of the plate covering the spherical reel has been made to fit the index finger (Fig. 3). Hence the operator should note that it requires but one complete turn from extreme side to side, rather than a number of short turns, in order to bring each individual lens into position, thus obtaining the full advantage of the automatic spring-stop. This likewise permits the refractionist to operate the Ski-optometer even though the room is in total darkness.

CONCAVE SPHERICAL LENSES

Another simple and exclusive Ski-optometer advantage worthy of note is the method employed in obtaining concave, spherical lenses. Instead of employing a battery of concave lenses similar to the convex battery previously described, the instrument’s operation is greatly simplified through the use of a neutralizing process.

In short, the Ski-optometer only contains two concave lenses to obtain its entire series—namely, a -6.D and a -12.D sphere (Fig. 3c)—first setting the pointer of the supplementary disk at -6. sphere, then setting the indicator of the spherical battery at +6.

Thus zero (or plano) is obtained, the plus neutralizing the minus.

By merely turning the plus or convex spherical reel inward, or toward the patient’s nose, the convex power is then decreased, naturally increasing the concave value or total minus lens power. For example, if the spherical indicator shows +5.D, when the -6D. lens is placed behind it, the lens value at the sight opening will be -1D (Fig. 6). If required, the refractionist may continue on this plan until only the -6D. lens remains.

Fig. 6—With this indicator of supplementary disk, set at -6.D. Sph. and spherical indicator at +5.D—lens value at sight opening is -1.D. Sph. This simple arrangement makes it possible to operate the Ski-optometer with but Single Reel for both plus and minus sphericals.

Should concave power stronger than -6D. be desired, by placing the pointer of the supplementary disk at -12D. Sph. and proceeding to neutralize as before, all the concave powers up to -12D. in quarters are similarly obtained. For the convenience of the operator, all minus or concave spherical powers are indicated in red; while plus, or convex powers, are indicated in white.

The instrument is also provided with an opaque or blank disk which is brought into position before the sight opening by setting the pointer of the supplementary disk at “shut” (Fig. 3c.)

Summing up, all plus and minus spherical powers have been attained from zero to 12D. in quarters, practically through the turn of the single reel—a simplicity of operation largely responsible for Ski-optometer supremacy.

CHAPTER II

CYLINDRICAL LENSES

IT IS commonly admitted that setting each trial-case cylindrical lens at a common axis is the most tedious part of refraction.

The automatic cylinder, one of the Ski-optometer’s latest and distinctly exclusive features, not only overcomes this annoyance but also avoids the need of individually transferring each cylindrical lens according to the varying strengths.

Fig. 7—Once you set the axis indicator as shown by dotted fingers, each cylindrical lens in the instrument automatically positions itself exactly at that axis, as indicated by the arrow.

By merely setting the Ski-optometer’s axis indicator (Fig. 7), each cylindrical lens in the instrument automatically positions itself, so that it will appear at the opening at the exact axis indicated.

This is readily accomplished by placing the thumb on the small knob, or handle of the axis indicator, drawing it outward so as to release it from spring tension. The indicator may then be set at any desired axis; and, on releasing the handle, every cylinder in the instrument becomes locked, making it impossible for any lens to appear at an axis other than the one specified by the indicator.

This insures the absolute accuracy of the axis of every cylinder as it appears before the patient’s eye. Subsequent shifting of the axis even to a single degree is impossible, although it is a common occurrence where trial-case lenses are employed.

OBTAINING CORRECT FOCUS

After setting the axis indicator, the only remaining move is to obtain the correct cylindrical strength or focus. This is readily accomplished by merely turning the Ski-optometer’s larger or extreme outer single reel, which contains concave cylindrical lenses from .25D to 2D in quarters (Fig. 8a). It should again be borne in mind that a downward turn increases concave cylinder power, while an upward turn decreases it. The operation of the cylinder reel is greatly facilitated by carefully noting position of thumb and index finger (Fig. 8). Thus accuracy of result, simplicity of operation and the saving of much valuable time is invariably assured.

Fig. 8A—Inner cog-wheel construction, showing arrangement of Ski-optometer cylinders. This simple construction assures accuracy and avoidance of the slightest shifting of axes.

As each cylinder appears before the patient’s eye, it simultaneously registers its focus at the indicator marked “CC CYL” shown in Fig. 8. Examinations of greater accuracy could not possibly be made than those obtained through the Ski-optometer, hence no refractionist should hesitate to employ it throughout an entire examination—wherever trial-case lenses are used.

The range of the Ski-optometer’s cylinder lens battery includes up to 2D. in quarters. An axis scale and a cell is located at the back of the instrument for insertion of an additional trial-case cylinder lens, when stronger cylindrical power is required. For example, if an additional -2D. cylinder is added, it will increase the range up to 4D. cylinder; or if twelfths are desired, a 0.12D. cylinder lens may be inserted. In this connection, it is interesting to note that considerable experimenting with twelfths in the Ski-optometer proved them to be needless, inasmuch as the instrument’s cylindrical lenses set directly next to the patient’s eyes overcome all possible loss of refraction, as explained in a later paragraph.

Fig. 8—Turn this Single Reel as shown by dotted finger to obtain cylindrical lenses, which simultaneously register their focus as they appear. Each lens also automatically positions itself at axis designated.

WHY CONCAVE CYLINDERS ARE USED EXCLUSIVELY

The Ski-optometer contains only concave cylinders, as it is universally admitted that convex cylinders are not essential for testing purposes.

In fact, concave cylinders should alone be used in making an examination, even where a complete trial-case is employed. To repeat one of the first rules of refraction: “As much plus or as little minus spherical power as patients will accept, combined with weakest minus cylinder, simplifies the work of refraction and insures accuracy without time-waste.”

After an examination with the Ski-optometer is completed, the total result of plus sphere and minus cylinder may be transposed if desired, though in most instances it is preferable to prescribe the exact findings indicated by the instrument. This will also avoid every possibility of error, eliminating responsibility where one is not familiar with transposition—since, after all, it is the duty of the optician to thoroughly understand that part of the work.

TRANSPOSITION OF LENSES

It is commonly understood that transposition of lenses is merely change of form, but not of value.

For example, a lens +1.00 sph. = -.50 cyl. axis 180° may be transposed to its equivalent, which is +.50 sph. = +.50 cyl. axis 90°. The accepted formula in this special instance is as follows: Algebraically add the two quantities for the new sphere, retain the power of the original cylinder, but change its sign and reverse its axis 90 degrees. Applying this rule, a lens +.75 sph. = -.25 cyl. axis 180°, is equivalent to +.50 sph. = +.25 cyl. axis 90°.

Similarly, a lens +1.00 sph. = -1.00 cyl. axis 180° is equivalent to +1.00 cyl. axis 90°.

One of the difficulties in transposing is in reversing the axis. In such cases, it is well to memorize the following simple rule:

To reverse the axis of any cylindrical lens containing three numerals—add the first two together and carry the last. For example, from 105 to 180 degrees, etc.:

105°

Add—one and “0” equals 1

Then carry the 5 = 15°

120°

Add—one and two equals 3

Then carry the 0 = 30°

130°

Add—three and one equals 4

Then carry the 0 = 40°

150°

Add—five and one equals 6

Then carry the 0 = 60°

165°

Add—six and one equals 7

Then carry the 5 = 75°

180°

Add—eight and one equals 9

Then carry the 0 = 90°

To transpose where there are but two numerals,

90° should be added.

In using the Ski-optometer, it is absolutely unnecessary to transpose the final result of an examination; merely write the prescription as instrument indicates. The idea that plus sphere combined with minus cylinder, or the reverse, is an incorrect method of writing a prescription, has long since been disproved.

CHAPTER III

HOW THE SKI-OPTOMETER ASSISTS

IN REFRACTION

THE construction of the Ski-optometer has now been fully explained, and the reader realizes that since the instrument contains all the lenses necessary in making an examination, greater operative facility is afforded through its use than where the trial-case lenses are employed.

The Ski-optometer is “an automatic trial-case” in the broadest sense of the term, wholly superseding the conventional trial-case. It should therefore be employed throughout an entire examination, wherever trial-case lenses were formerly used. To fully realize its labor saving value in obtaining accurate examination results, it is only necessary to recall the tedious method of individually handling and transferring each lens from the trial-case to the trial-frame, watching the stamped number on each lens handle, wiping each lens and in the case of cylindrical lenses setting each one at a designated axis—all being needless steps where the Ski-optometer is employed.

THE USE OF THE SKI-OPTOMETER IN SKIOSCOPY

In skioscopy, the Ski-optometer offers the refractionist assistance of the most valuable character.

For example, assuming that extreme motion in the opposite direction with plane or concave mirror is obtained with a +1.25D. spherical lens before the patient’s eye; by quickly turning the Ski-optometer’s single reel until the two white zeros again appear, +2.50D is secured, as explained in the previous chapter. If this continues to give too much “against motion,” the lens power should be quickly increased to +3.75 or +5.00D if necessary (Fig. 4). Should the latter reveal a shadow in the reversed direction, the refractionist is assured that it is the weakest lens that will cause its neutralization. Practically but few lenses have been used to obtain the final result proving the instrument’s importance and time-saving value in skioscopy, and demonstrating the simplicity with which tedious transference of trial-case lenses is avoided.

Furthermore, it should be noted that where the Ski-optometer is used in skioscopy, it is not necessary to remove the retinoscope from the eye or to constantly locate a new reflex with each lens change. This permits a direct comparison of the final lens and eliminates the usual difficulty in mastering skioscopy. The chief cause of this difficulty is due to the fact that the transferring of the trial-case lenses makes it practically impossible for the student to determine whether the previous lens caused more “with” or “against” motion.

Fig. 9—The Woolf ophthalmic bracket. A convenient and portable accessory in skioscopy and muscle testing; can be used with or without Greek cross.

Where the indirect method is employed in skioscopy, best results are secured through the use of the Woolf ophthalmic bracket and concentrated filament lamp, together with an iris diaphragm chimney. The latter permits the reduction or increase of the amount of light entering the eye, as it is agreed that a large pupil requires less light, a small pupil requiring more light. The bracket referred to permits the operator to swing the light into any desired position (Fig. 9), while the iris diaphragm chimney serves as a shutter. This apparatus may also be employed for muscle testing, as described in a subsequent paragraph.

A SIMPLIFIED SKIOSCOPIC METHOD

In using the Ski-optometer, instead of working forty inches away from the patient in skioscopy and deducting 1.D., the refractionist will find it more convenient to work at a twenty inch distance, deducting 2.D. This working distance may be accurately measured and maintained by using the reading rod accompanying the instrument. Instead of deducting 2.D. from the total findings, however, it is preferable to insert a +2.D. trial-case lens in the rear cell of the instrument directly next to the patient’s eye. After determining the weakest lens required to neutralize the shadow in both meridians, the additional +2.D. lens should be removed and the total result of the examination read from the instrument’s register.

To illustrate a case in skioscopy where spherical lenses are employed to correct both meridians, assume that the vertical shadow requires a +1.25D lens to cause its reversal, while the horizontal requires +2.00D. Employment of the customary diagram, illustrated in Fig. 10, would show the patient required +1.25 sph. = +.75 cyl. axis 90°, which when transposed is equivalent to +2.00 sph. = -.75 cyl. axis 180°.

Fig. 10—Where spherical lenses are employed in skioscopy, above indicates patient requires

+1.25 Sph. = +.75 Cyl. Axis 90°

or   +2 Sph. = -.75 Cyl. Axis 180°

It should be noted that the total spherical power is +2.00D, as the Ski-optometer’s register shows, while the difference between the two meridians is 75, which is the required strength of the cylinder. By then turning the cylinder reel to .75, and setting the axis indicator at 180° (because by using minus cylinders, the axis must be reversed) the patient should read the test-type with ease if the skioscopic findings are correct. Thus with the Ski-optometer, it is not even necessary to learn transposition, since the instrument automatically accomplishes the work, avoiding all possibility of error.

EMPLOYING SPHERES AND CYLINDERS

IN SKIOSCOPY

Another commonly used objective method may be employed with even greater facility through the combined use of both the Ski-optometer’s spherical and cylindrical lenses. As previously suggested, insert the +2.00 spherical trial-case lens in the rear of the instrument, working at a twenty inch distance, then proceed to correct the strongest meridian first.

It was assumed that it required a +2.00 spherical to neutralize the strongest, or horizontal meridian, as shown in Fig. 10. The refractionist should then set the axis indicator therewith, which is the axis of the cylinder, or 180°.

It is then merely a matter of increasing the Ski-optometer’s cylindrical lens power until the reversal of the shadow in the weakest meridian is determined. Assuming this proves to be -.75 cylinder, axis 180°, the patient’s complete prescription +2.00 sph. = -.75 cyl. axis 180°, would be registered in the Ski-optometer without any further lens change other than the removal of the +2.00 working distance lens.

However, regardless of the method employed, the Ski-optometer greatly simplifies skioscopy. In fact, the instrument was originally intended to simplify retinoscopy or skioscopy, as the subject should be termed, the name “Ski-optometer” having been derived from the latter.

USE OF THE SKI-OPTOMETER

IN SUBJECTIVE TESTING

In subjective refraction, especially where the “better or worse” query must be decided by the patient, it is commonly understood that the refractionist is compelled to first increase and then decrease a quarter of a diopter before the final lens is decided. With the Ski-optometer, the usual three final changes are made in far less time than it takes to make even one lens change from trial-case to trial-frame.

For example:

Assuming, with a +1.25D spherical lens before the patient’s right eye, he remarks that he “sees better” with a +1.D. while +.75D is not as satisfactory. The refractionist can then quickly return to +1.D., simply turning the Ski-optometer’s single reel outward to increase, or backward to decrease, the lens strength. So rapidly have these lens changes been made, that the patient quickly sees the difference of even a quarter diopter, and quickly replies, “better” or “worse.”

This is made possible because the eye does not “accommodate” as quickly as the lens change made with the Ski-optometer. It should also be noted that the eye receives an image on its retina within one-sixteenth of a second; otherwise, the patient is forced to accommodate, making it difficult to see the difference of even a quarter diopter. On the other hand, in transferring trial-case lenses, with its slow, tedious procedure, the patient, being unable to detect the slight difference of only a quarter diopter, unhesitatingly replies, “no difference,” merely because they are compelled to accommodate.

A SIMPLIFIED SUBJECTIVE METHOD

The following simplified method of procedure is suggested for subjective testing with the Ski-optometer, although as previously explained, the refractionist may employ his customary method, overcoming the annoyance of transferring trial-case lenses and the setting of each cylinder individually. The Ski-optometer has been constructed and based upon the golden rule of refraction: “As much plus or as little minus spherical, combined with as little minus cylinder power as the patient accepts.”

By applying this rule as in the above method and starting with +5.D. spherical, watching the two zeros (Fig. 4) and rapidly reducing +1.25D each time, we will assume that +1.25D gives 20/30 vision; as a final result +1.D. will possibly give 20/25 vision.

The patient’s attention should next be directed to the most visible line of type, preferably concentrating on the letter “E” or the clock dial chart—either of which will assist in determining any possible astigmatism. Since the Ski-optometer contains concave cylinders exclusively, the next move should be the setting of its axis indicator at 180°, commonly understood as “with the rule.” One should then proceed to determine the cylinder lens strength by turning the reel containing the cylindrical lenses (Fig. 8). Should the patient’s vision fail to improve after the -.50D. cylinder axis 180° has been employed, the refractionist, in seeking an improvement, should then slowly move the axis indicator through its entire arc.

With the cylinder added, regardless of axis, poor vision might indicate the absence of astigmatism. If astigmatism exists, vision will usually show signs of improvement at some point, indicating the approximate axis. Once the latter is ascertained, the refractionist may readily turn the Ski-optometer’s cylinder reel and obtain the correct cylinder lens strength, after which the axis indicator should be moved in either direction in order to obtain the best possible vision for the patient.

The refractionist should always aim to obtain normal (or 20/20) vision with the weakest concave cylinder, combined with the strongest plus sphere, or weakest minus sphere.

PROCEDURE FOR USING MINUS CYLINDERS

EXCLUSIVELY

For the benefit of those who have never used minus cylinders exclusively in making their examinations, we will assume that the patient requires O.U. +1.D sph. = -1D cyl. axis 180° for final correction; the latter, in its transposed form, being equivalent to +1.D. cylinder axis 90°. Unquestionably the best method is the one that requires the least number of lens changes to secure the final result.

To obtain this, the following order of lens change should be made: First, +1.D. sphere is finally determined and allowed to remain in place. Concave cylinders are then employed in quarters until the final results of +1.D. spherical, combined with -1.D. cylinder axis 180° is secured. This necessitates the change of but four cylindrical lenses as shown in routine “A” as follows:

ROUTINE “A”

ROUTINE “B”

(Made with minus cylinder)

(Made with plus cylinder)

Sph.

+1.D.

Cyl.

Axis

Sph.

+1.D.

Cyl.

Axis

Step 1

+1.D. =

-.25 ax.

180°

equal to

+.75 =

+.25 ax.

90°

Step 2

+1.D. =

-.50 ax.

180°

equal to

+.50 =

+.50 ax.

90°

Step 3

+1.D. =

-.75 ax.

180°

equal to

+.25 =

+.75 ax.

90°

Step 4

+1.D. =

-1 ax.

180°

equal to

0

+1 ax.

90°

In brief the method of using minus cylinders exclusively in an examination, as explained in routine “A”, necessitates the change of the cylinder lenses only after the strongest plus sphere is secured.

On the other hand, notwithstanding innumerable other methods where plus cylinders are used, routine “B” shows that the best spherical lens strength the patient will accept, is also first determined. Then both spheres and cylinders are changed in their regular order by gradually building up in routine, by increasing plus cylinder and next decreasing sphere, a quarter diopter each time, until the final result is secured.

While it is conceded that both routine “A” and “B” are of themselves simplified methods, by comparing routine “A” where minus cylinders are used with routine “B” where plus cylinders are used in their corresponding steps, the refractionist will note by comparison that one is the exact equivalent and transposition of the other. Where plus cylinders are employed, eight lens changes are made before final results are secured; while but four lens changes are necessary where minus cylinders are used.

The refractionist should also note by comparison that the use of minus cylinders reduces focus of the plus sphere, but only in the meridian of the axis. It has not made the patient myopic. Furthermore, a plus cylinder will bring the focal rays forward, while minus cylinders throw them backward toward the retina.

This is but another reason for the exclusive use of minus cylinders in refraction.

The method of using minus cylinders exclusively in an examination, necessitates the change of the cylinder lenses only. On the other hand, the method of using plus cylinders makes it necessary to change spheres and cylinders in routine.

In brief, since using the minus cylinder is merely a matter of mathematical optics, their use even in a trial-case examination is strongly urged.

The maximum value of the Ski-optometer is fully realized only when the advantages of using minus cylinders exclusively in every examination is clearly understood.

CONSTANT ATTENTION NOT REQUIRED

With the Ski-optometer, when the examination is completed, the sum-total of final results—whether spherical, cylinder, axis, or all combined—are automatically indicated or registered ready to write the prescription. Until then, the foci of the various lenses that may be employed are of no importance.

In short, in using the Ski-optometer, it is not necessary to constantly watch the registrations during examinations. The automatic operation of the instrument is an exclusive feature, so that the refractionist should unhesitatingly employ it. Hence, by eliminating the perpetual watch on the lenses in use, the refractionist is enabled to give his undivided attention to the patient rather than to the trial lenses.

Where a special dark-room is used for skioscopic work, an additional wall bracket or floor stand will necessitate only the removal of the instrument itself. This enables the refractionist to use the Ski-optometer for subjective or objective work, without disturbing the patient’s correction.

CHAPTER IV

IMPORTANT POINTS IN CONNECTION WITH

THE USE OF THE SKI-OPTOMETER

THE Ski-optometer is equipped with an adjustable head-rest, permitting its lenses to be brought as close as possible to the eye without touching the patient’s lashes, a matter of importance in every examination.

Fig. 11—The nasal lines of the Ski-optometer fit the contour of face with mask-like perfection, patient remaining in comfortable position.

ELIMINATION OF TRIAL-FRAME DISCOMFORT

Where the Ski-optometer is correctly fitted to the face, the patient invariably remains in a comfortable position (Fig. 11). The instrument is shaped to fit the face like a mask, so that even with a pupillary distance of but 50 m/m (that of a child) there still remains, without pinching, ample room for the widest nose of an adult.

Before making an examination, the correct pupillary distance should always be obtained by drawing an imaginary vertical line downward through the center of each eye from the 90° point on the Ski-optometer axis scale. The pupillary distance will then register in millimeters on the scale of measurements for each eye separately. If the Ski-optometer is correctly adjusted, the patient is securely held in position, the cumbersome trial-frame being entirely eliminated.

RIGIDITY OF CONSTRUCTION

Illustration on following page (Fig. 11a) shows the reinforced double bearing arms which hold the Ski-optometer lens batteries at two points. This eliminates possibility of the instrument getting out of alignment, and prevents wabbling or loose working parts.

The broad horizontal slides shown in the cut, move in and out independently so that the pupillary distance is obtained for each eye separately by turning the pinioned handle on either side of the instrument. The scale denotes in millimeters the P.D. from the median line of the nose outward, the total of both scales being the patient’s pupillary distance.

Fig. 11a also serves to show the staunch construction of the base of the Ski-optometer.

Fig. 11a—Showing staunch construction of Ski-optometer base.

HOW TO PLACE THE SKI-OPTOMETER IN POSITION

The patient should be placed in a comfortable position with “chin up,” as though looking at a distant object. The instrument should then be raised or lowered by the adjustable ratchet wheel of the bracket. The wall bracket gives best results when suspended from the wall, back of the patient, as shown on page 135. This bracket should be placed about ten inches above the head of the average patient. When the Ski-optometer is placed in position for use, its lower edge will barely touch the patient’s cheeks. It is sometimes advisable to request the patient to lightly press toward the face the horizontal bar supporting the instrument. Particularly good results are secured where a chair with a head-rest is employed in conjunction with the Ski-optometer. (See illustration of Model Refraction Room, Page 112).

CLEANING THE LENSES

The time-waste of perpetually cleaning lenses is overcome where the Ski-optometer is employed. For the convenience of the operator and protection of Ski-optometer lenses, the latter are concealed in a dust-proof cell, overcoming all dust and finger-print annoyances. When not in use, the instrument should be covered with the standardized hood forming part of the equipment.

The instrument should not be taken apart under any circumstances. To clean its lenses, not a single screw need be removed, as the lenses of each disk may be cleaned individually through the opening of the other disks. These openings are conveniently indicated by the white zeros (Fig. 2). The Ski-optometer contains but eleven spherical and eight cylindrical lenses on each side, so that the actual work of cleaning should not require over ten minutes at the most, cleaning the lenses every other week proving quite sufficient.

ACCURACY ASSURED IN EVERY TEST

Loss of refraction is completely eliminated through the use of the Ski-optometer. The most casual examination of the trial-frame or any other instrument shows that the construction necessitates the placing of the spherical lens next to the eye with the cylinder lens outermost—a serious fault wholly overcome in the Ski-optometer.

Not only do the cylindrical lenses of the Ski-optometer set directly next to the patient’s eye, thus overcoming any possible loss of refraction, but the strong spherical lenses of the supplementary disk are set directly next to the cylinder. There is apparently but a hair’s distance between these lenses; the two disks containing the spherical lenses of the Ski-optometer likewise setting close together.

In a word, the Ski-optometer’s cylinder lenses set directly next to the patient’s eye, followed by the stronger sphericals, so that the weakest spherical or +.25 (the lens of least importance) sets farthest away. This is 3½ m/m closer than any trial-frame manufactured, however, and at least 10 m/m closer than any other instrument—another reason for implicitly relying on the Ski-optometer for uniformly accurate results.

BUILT TO LAST A LIFETIME

Fig. 12—(A. and B.)—This unique, patented split-spring device of screwless construction, securely holds all movable parts. In case of repair, they may be removed with the blade of a knife.

The Ski-optometer is built on the plan of ¹/₁₀₀₀″, insuring absolute rigidity and accuracy and a lifetime of endurance. Particular and detailed attention has been given to the novel means of eliminating screws which either bind, create friction or continually work loose, causing false indications of findings on scales of measurements; hence correct and accurate indications are insured in the Ski-optometer by means of a split-spring washer construction similar to that of an automobile tire’s detachable rim (Fig. 12).

This patented spring washer construction securely holds the phorometer lenses, the rotary prism and the revolving cylinder lens cells.

Whenever necessary, or in case of repair, these parts may be readily removed with the blade of a knife.

CHAPTER V

CONDENSED PROCEDURE FOR MAKING

SPHERE AND CYLINDER TEST

WITH THE SKI-OPTOMETER

NOTWITHSTANDING various methods employed, for both subjective and objective refraction, the following synopsis of the previous chapters will unquestionably prove most valuable to the busy refractionist, enabling him to make error-proof examinations in practically every case without resorting to the transference of trial-case sphere or cylinder lenses. A careful reading of chapters one and two should be made however, so that one may gain an understanding as to how spheres and cylinders are obtained with the Ski-optometer.

SUBJECTIVE DISTANCE TEST

1st—Place Ski-optometer in position, employing spirit level, thus maintaining instrument’s horizontal balance.

2nd—Adjust the pupillary distance for each eye individually, by drawing an imaginary vertical line downward through the center of each eye from the 90° point on the Ski-optometer’s axis scale. The opaque disk should be placed before the patient’s left eye by setting the supplementary disk handle at “shut.”

3rd—The Ski-optometer lens battery before the patient’s right eye should be set at “open” (figure 2), whereupon the first turn of spherical lens battery toward the nasal side places a +6.D sphere in position. This should blur vision of average patient.

4th—It is now only necessary to remember that an outward turn toward temporal side of the instrument increases plus sphere power, while a nasal turn decreases it. Therefore continue to reduce convex spherical lens power until the large letter “E” on the distant test card is clear. Then request patient to read as far down as possible,—a rapid turn of a quarter diopter being readily accomplished with the Ski-optometer (Fig. 4).

5th—In the event of working down to “zero” with spheres, the supplementary disk handle or indicator should next be set at -6.D sphere, while the spherical reel should be turned toward the nasal side—thus building up on minus spheres (Fig. 6). In short, the strongest plus sphere or weakest minus sphere should always be determined before employing cylinders.

6th—With the best spherical lens that the patient will accept left in place, direct attention to the letter E or F in the lowest line of type the patient can see on the distant test letter chart. Then set axis indicator at 180° (Fig. 7).

7th—Next increase concave cylinder power until vision is improved. If vision is not improved after increasing cylinder strength to -.50 axis 180°, merely reverse the axis to 90°. If vision is improved, cylinder lens strength should be increased. If not, it should be decreased (Fig. 8).

8th—Slowly move axis indicator through entire arc of axis, thus locating best possible axis (Fig. 7).

9th—After sphere and cylinder test of right eye has been made, place supplementary disk handle at “shut.” Then repeat procedure in testing left eye.

10th—After completing examination for each eye separately, then, with both of the patient’s eyes open, direct attention to lowest line of type he can see, concentrating on the E or F, simultaneously increasing or decreasing spherical power before both eyes. The refractionist merely recalls that by turning the Ski-optometer’s single reel toward the temporal side, convex spherical power is increased, by turning toward the nasal side for either eye, spherical power is decreased. Cylinder lens strength may be changed in a like manner before both eyes simultaneously.

11th—After making the distance test, then only is it necessary to copy the result of the examination as recorded by the Ski-optometer.

SUBJECTIVE READING TEST

Tilt Ski-optometer forward in making reading test. The wide groove in the horizontal bar supporting the instrument, permits it to be slightly tilted.

12th—Place Ski-optometer reading rod in position with card at about 14 inches. Close off one eye. Direct patient’s attention to the name “Benjamin” printed at top of card.

13th—Leave cylinder lens in place. Proceed as in distance test with +6.D sphere, fogging down until the first word “laugh” on the reading card, in line 75M, is perfectly clear, this being slightly smaller than the average newspaper type.

14th—After completion of examination for each eye separately, then with both eyes direct patient’s attention to word “laugh.” Move reading card in or out a few inches either side of 14 inch mark. This will determine any possibility of an over-correction. Then record prescription just as Ski-optometer indicates. For a detailed description of above, as well as for objective testing with the Ski-optometer, read chapter three.

CHAPTER VI

MUSCULAR IMBALANCE

THE purpose of the present chapter is to acquaint the refractionist with the operation of the Ski-optometer as “a scientific instrument for muscle testing”—the subject being treated as briefly and comprehensively as is practicable.

As the reader progresses in the subject of muscular anomalies, he may carry his work to as high a plane as desired, increasing his professional usefulness to an enviable degree.

Through the use of the Ski-optometer, muscle testing may be accurately accomplished in less time than a description of the operation requires. Furthermore, tedious examinations may be wholly overcome through the discontinuance of the consecutive transference of the various degrees of prisms from the trial-case. In fact, the latter method has long been quite obsolete, owing to the possibility of inaccuracy. The muscle action of the eye is usually quicker than the result sought through the use of trial-case prisms; hence muscle testing with the Ski-optometer is accomplished with far greater rapidity and accuracy, thus making the instrument an invaluable appliance in every examination.

THE ACTION OF PRISMS

Students in refraction—and one may still be a student after years of refracting—are sometimes puzzled as to just what a prism does when placed before an eye. They refer to every available volume and are often confused between ductions and phorias, finally dropping the subject as an unsolvable problem. In view of this fact, it is suggested that the refractionist should read the present volume with the actual instrument before him.

Before proceeding, one should first understand the effect of a prism and what it accomplishes. To determine this, close one eye, looking at some small, fixed object; at the same time, hold a ten degree prism base in before the open eye, noting displacement of the object. This will clearly show that the eye behind the prism turns toward the prism apex.

To carry the experiment further, the following test may be employed on a patient. Covering one eye, direct his attention to a fixed object, placing the ten degree prism before the eye, but far enough away to see the patient’s eye behind it. As the prism is brought in to the line of vision, it will be seen that the eye turns towards the apex of the prism. When the prism is removed, the eye returns to its normal position.

Similar experiments enable the refractionist to make the most practical use of treating phorias and ductions, as well as to comprehend all other technical work.

Fig. 13—An important part of the equipment for muscular work.

THE PHOROMETER

As previously stated, it is practically impossible to accurately diagnose a case of muscular imbalance with trial-case prisms. For this reason the phorometer forms an important part of the equipment for muscle testing in the Ski-optometer, having proven both rapid and accurate. It consists of two five-degree prisms with bases opposite, each reflecting an object toward the apex or thin edge. The patient whose attention is directed to the usual muscle-testing spot of light, will see two spots.

Aside from the instrument itself, and in further explanation of the phorometer’s principle and construction, when two five-degree prisms are placed together so that their bases are directly opposite, they naturally neutralize; when their bases are together, their strength is doubled. Thus while the prisms of the phorometer are rotating, they give prism values from plano to ten degrees, the same being indicated by the pointer on the phorometer’s scale of measurements.

As a guide in dark-room testing, it should be noted that the handle of the phorometer in a vertical position is an indication that the vertical muscles are being tested; if horizontal, the horizontal muscles are undergoing the test.

THE MADDOX ROD

The Maddox rod (Fig. 14) consists of a number of red or white rods, which cause a corresponding colored streak to be seen by the patient. This rod is placed most conveniently on the instrument, being provided with independent stops for accurately setting the rods at 90 or 180 degree positions. The Maddox rod has proven of valuable assistance in detecting muscular defects, particularly when used in conjunction with the phorometer. Thus employed, it enables the patient to determine when the streak seen with one eye crosses through the muscle-testing spot-light observable by the other eye, as hereafter described.

Fig. 14—The Maddox Rod,

a valuable aid in making muscular tests.

PROCEDURE FOR MAKING THE MUSCLE TEST

The Ski-optometer should be equipped with two Maddox rods, one red and one white. Their combined use is of the utmost importance since they assist in accurately determining cyclophoria and its degree of tortion as designated on the degree scale, and fully described in a later chapter.

When the Maddox rods are placed in a vertical position, it is an indication that the vertical muscles are being tested; when placed horizontally, the horizontal muscles are being tested. It should be particularly noted that the streaks of light observable through the Maddox rods always appear at right angles to the position in which they lie.

The Ski-optometer should be placed in a comfortable position before the patient’s face with the brow-rest and pupillary distance adjusted to their respective requirements. The instrument should be levelled so that the bubble of the spirit level lies evenly between its two lines, thus insuring horizontal balance. The muscle test light should be employed at an approximate distance of twenty feet on a plane with the patient’s head. Best results in muscle testing are secured through the use of the Woolf ophthalmic bracket, with iris diaphragm chimney and a specially adapted concentrated filament electric lamp (Fig. 9). This gives a brilliant illumination which is particularly essential. The test for error of refraction should be made in the usual manner, using the spherical and cylindrical lenses contained in the Ski-optometer, thus obviating the transference of trial-case lenses and the use of a cumbersome trial-frame. The time-saving thus effected enables the refractionist to include a muscle test in every examination and without tiring the patient—a consideration of the utmost importance.

BINOCULAR AND MONOCULAR TEST

The test for muscular imbalance may be divided in two parts. First, binocular test, or combined muscle test of the two eyes; second, monocular test, or muscle test of each eye separately. The latter does not signify the shutting out of vision or closing off of either eye, since muscular imbalance can only be determined when both eyes are open. These two tests are fully explained in the following chapter.

CHAPTER VII

THE BINOCULAR MUSCLE TEST

MADE WITH THE MADDOX ROD

AND PHOROMETER

DIRECTING the patient’s attention to the usual muscle testing spot of light, the red Maddox rod should be placed in operative position before the eye, with the single white line or indicator on red zero (Fig. 15). The rods now lie in a vertical position.

Fig. 15—The Maddox rods placed vertically denote test for right or left hyperphoria, causing a horizontal streak to be seen by patient.

The pointer of the phorometer should likewise be set on the neutral line of the red scale, causing the handle to point upward (Fig. 16). A distance point of light and a red streak laying in a horizontal position should now be seen by the patient.

Fig. 16—The phorometer handle placed vertically, denotes vertical muscles are undergoing test for right or left hyperphoria—as indicated by “R. H.” or “L. H.”

Instead of memorizing a vast number of rules essential where trial case prisms are employed for testing ocular muscles, the pointer of the phorometer indicates not only the degree on the red scale, but the presence of right hyperphoria (R. H.) or left hyperphoria (L. H.).

Fig. 17—The horizontal streak caused by Maddox rod bisecting muscle testing spot-light for vertical imbalance, as patient should see it.

Assuming that the patient finds that the streak cuts through the point of light, the refractionist instantly notes the absence of hyperphoria. Should the point of light and the red streak not bisect, prism power must be added by rotating the phorometer’s handle to a position that will cause the streak to cut through the light (Fig. 17). While testing for hyperphoria, the red scale should alone be employed, the white scale being totally ignored.

Fig. 18—The Maddox rods placed horizontally test esophoria or exophoria, causing a vertical streak to be seen by the patient.

ESOPHORIA AND EXOPHORIA

The next step is to set the white lines of the red Maddox rod either at white zero, or 180° line, with the rods in a horizontal position (Fig. 18) and the phorometer on the white neutral line, with handle horizontal (Fig. 19), thus making the test for esophoria or exophoria, technically known as lateral deviations.

The red streak will now be seen in a vertical position. Should it bisect the spot of light, it would show that no lateral imbalance exists. Should it not bisect, the existence of either esophoria or exophoria is proven, necessitating the turning of the phorometer handle. Should the refractionist rotate the handle in a direction opposing that of the existing imbalance, the light will be taken further away from the streak, indicating that the rotation of the prisms should be reversed.

Fig. 19—The phorometer handle placed horizontally denotes horizontal muscles are undergoing test for esophoria or exophoria indicated by “Es.” or “Ex.”

At the point of bisection (Fig. 20), the phorometer will indicate on the white scale whether the case is esophoria or exophoria and to what amount. In testing esophoria (ES) or exophoria (EX), the white scale is alone employed, no attention being given to the red scale.

Fig. 20—The vertical streak bisecting muscle testing spot-light for horizontal imbalance, as patient should see it.

MAKING MUSCLE TEST BEFORE AND

AFTER OPTICAL CORRECTION

It is considered best to make the binocular test before regular refraction is made, making note of the findings; and again repeating the test after the full optical correction has been placed before the patient’s eye. This enables the refractionist to definitely determine whether the correction has benefited or aggravated the muscles. Furthermore, by making the muscle test before and after the optical correction, a starting point in an examination is frequently attained. For example, where the phorometer indicates esophoria it is usually associated with hyperopia, whereas exophoria is usually associated with myopia, thus serving as a clue for the optical correction.

Assuming for example that the binocular muscle test shows six degrees of esophoria without the optical correction, and with it but four degrees, it is readily seen that the imbalance has been benefited by the optical correction. Under such conditions it is safe to believe that the optical correction will continue to benefit as the patient advances in years, tending to overcome muscular defect.

WHEN TO CONSIDER CORRECTION

OF MUSCULAR IMBALANCE

In correcting an imbalance, it is also a good plan to adhere to the following rule: In case of hyperphoria, either right or left, consider for further correction only those cases that show one degree or more. In exophoria, those showing three degrees or more. In esophoria, correct those showing five degrees or more, except in children, where correction should be made in cases showing an excess of 3° of esophoria. These rules are naturally subject to variation according to the patient’s refraction and age, but they are generally accepted as safe.

FOUR METHODS FOR CORRECTION

OF MUSCULAR IMBALANCE

There are four distinct methods for correcting muscular imbalance, each of which should be carried out in the following routine:

1. Optical correction made with spheres or cylinders, or a combination of both.

2. Muscular exercising or “ocular gymnastics.” This is accomplished on the same principle as the employment of other forms of exercises, or calisthenics.

3. The use of Prisms: When the second method fails, prisms are supplied, with base of prism before the weak muscle, for rest only.

4. Operation: If the above three methods, as outlined in the following chapters, have been carefully investigated, nothing remains but a tetonomy or advancement, or other operative means for relief and satisfaction to the patient.

THE ROTARY PRISM

The rotary prism of the Ski-optometer, (Fig. 21) consists of a prism unit, having a total equivalent of thirty degrees. It is composed of two fifteen-degree prisms, back to back, so that the turn of its pinion or handle causes each of its lenses to revolve, one on the other. When its bases are opposite, they neutralize; when directly together, they give a total value of thirty degrees. While revolving from zero to maximum strength, they give prism values which are indicated on the scale of measurements, the red line denoting the total prism equivalent.

Fig. 21—Turning rotary prism’s pinioned handle gives prism value from zero to 30° as indicated by prism’s red line indicator.

It is obviously essential to know where the base of the rotary prism is located. Therefore if prism in or out is desired, the zero graduations should be placed vertically and the red line or indicator set at the upper zero (Fig. 21).

A rotation inward to 10 would give a prism equivalent of ten degrees, base in. A rotation from zero to 10 outward would give a prism equivalent of ten degrees, base out, etc. With zero graduations horizontal and the red line or indicator set therewith, a rotation upward to ten on the scale would give a prism equivalent of ten degrees, base up. A rotation from zero downward to 10 would give a prism equivalent of ten degrees, base down.

An understanding of the foregoing will show that a rotation of the red line, or indicator, will give prism value from zero to 30, with base up, down, in or out.

USE OF THE ROTARY PRISM IN

BINOCULAR MUSCLE TESTS

Should a case be one of esophoria, exceeding the ten degree range of the phorometer, the rotary prism should be brought into operative position with cypher (0) graduations vertical (Fig. 21), while the red line or indicator should be set at 10 on the outer or temporal scale. The phorometer’s indicator should again be set on the center or neutral line on the white scale. The rotary prism will then add ten degrees to the esophoria reading indicated on the phorometer.

Should the case be one of exophoria, exceeding ten degrees, the indicator should be set at ten degrees upon the inner or nasal scale and the indicator of the phorometer should then be set at the white center or neutral line, as in the previous test. Should prism power ever be required to supplement the phorometer in hyperphoria, the rotary prism should be employed with zero graduations horizontal, and the red line or indicator set at ten degrees on upper or lower scale, as required.

CHAPTER VIII

THE MONOCULAR DUCTION MUSCLE TEST

MADE WITH BOTH ROTARY PRISMS

WHILE the previously described binocular muscle test made with the phorometer and Maddox rod, only determines the existence and amount of esophoria, exophoria, and hyperphoria, neither the faulty nor the deviating muscle is located, hence a monocular muscle test is essential in order to determine whether the muscles of the right or left eye are faulty. Furthermore, an imbalance may possibly be due to either a faulty muscular poise, or lack of nerve force in one or both eyes. A “duction test” should accordingly be made of each muscle of each eye separately, followed by a comparison of the muscular pull of both eyes collectively.

These tests are commonly termed adduction, abduction, superduction and subduction, and are defined in the order named. They include tests of the vertical and horizontal muscles of each eye, made individually by means of the rotary prisms, each being placed before the eye undergoing the test.

LOCATING THE FAULTY MUSCLE

The phorometer and the Maddox rod should be removed from operative position, discontinuing the use of the muscle-testing spot-light, employed in the previously described binocular test. The optical correction, if one is required, should be left in place, while the patient’s attention should be directed, with both eyes open, to the largest letter on the distant test chart; or if preferable, the Greek cross in the Woolf ophthalmic, chimney may be used. Either one, however, should be located on a plane with the patient’s head. As a guide for the operator, it might be well to remember that when the handle of the rotary prism is in a horizontal position, the lateral or horizontal muscles are being tested. On the other hand, when the handle is in a vertical position, the vertical muscles are undergoing the test.

ADDUCTION

Adduction, or relative convergence, is the power of the internal muscles to turn the eyes inward; prism power base out and apex in, is employed.

Fig. 22—To test adduction, base out is required. Rotary prism’s line or indicator should be rotated from zero outwardly.

To test abduction, base in is required. Indicator should be rotated inwardly from zero.

To test adduction of the patient’s right eye, the rotary prism should be placed in position before the right eye, the red line or prism indicator being registered at zero upon the prism upper scale. The two cyphers (0) should be placed in a vertical position with the handle pointed horizontally (Fig. 21). The rotary prism should then be rotated so that its red line or indicator is rotated outward from zero until the large letter—preferably the largest letter, which is usually “E”—on the distance test-type or the Greek cross previously referred to, first appears to double in the horizontal plane. The reading on the scale of measurements should accordingly be noted. This test should be repeated several times, constantly striving for the highest prism power that the patient will accept without producing diplopia. The prism equivalent thus obtained will indicate the right adduction and should be so recorded, as designated in Fig. 24. The amount of adduction ranges from 6 to 28, prism diopters, the normal average being 24.

ABDUCTION

Abduction is the relative power of the external muscles to turn the eyes outward. Prism power base in and apex out is employed. To determine abduction, or the amount of divergence of the external rectus muscle of the right eye, prism power with base in or toward the nasal side should be employed. The rotary prism will therefore remain in the same relative position as in making the adduction test (Fig. 22), with the two cyphers (0) or zero graduations vertical, but the indicator or red line should be rotated inward from zero, or towards the patient’s nose.

With the patient’s attention again directed to the large letter “E,” or the Greek cross, this inward rotation should be continued until diplopia or double vision occurs. Like the former, this test should be repeated several times, the refractionist continuing to strive for the highest prism power which the eye will accept. This will indicate abduction of the right eye and should be so recorded as designated in Fig. 24. The amount of abduction ranges from 3 to 10 prism diopters. The normal average is 8.

The ratio of adduction to abduction is normally rated at about three to one. In other words, it is conceded that the power of the eye to converge is normally three times as great as its power to diverge, the usual measurements being eight to twenty-four respectively. While applicable in most instances, this may vary in different cases.

SUPERDUCTION

Superduction, sometimes termed sursumduction, is the relative power of the superior recti to turn the eyes upward. Prism power base down and apex up is employed. To test superduction, the rotary prism should be placed in position with the two cyphers lying horizontally, with the handle pointed vertically (Fig. 23). The patient’s attention should again be directed to the large letter “E”, and the indicator or red line should be rotated downward from zero. The highest prism power that the patient will accept before the object appears to double in the vertical plane will indicate the degree of right superduction. This should be recorded accordingly. Conditions of this kind do not usually exceed two or three degrees. The test, however, should be repeated several times before the final result is recorded, as indicated in Fig. 24. The amount of superduction ranges from 1 to 4 prism diopters. The normal average is 2.

Fig. 23—To test superduction, base down is required. Rotary prism’s line or indicator should be rotated downward from zero.

To test subduction, base up is required. Indicator should be rotated upward from zero.

SUBDUCTION

Subduction, sometimes termed infraduction or deorsumduction, is the relative power of the inferior recti to turn the eyes downward. Prism power base up and apex down is employed. To test subduction, the rotary prism should be operated with zero graduations placed horizontally, as in the superduction test (Fig. 23), but the indicator should be slowly rotated in the reverse direction, or upward from zero. With the patient’s attention again directed to the large letter “E,” or the Greek cross, the strongest degree prism thus secured without diplopia will indicate the right subduction. The amount of subduction ranges from 1 to 4 prism diopters. The normal average is 2.

Any difference between superduction and subduction, usually denoting the existence of hyperphoria, should be given careful consideration.

PROCEDURE FOR MONOCULAR MUSCLE TESTING

As previously explained, after a duction test of each of the four muscles of the right eye, the rotary prism before that eye should be placed out of position and the procedure for adduction, abduction, superduction and subduction repeated by means of the rotary prism before the left eye. In case of an existing imbalance, after testing the muscle of both right and left eyes, the refractionist can quickly determine which muscle or muscles may be lacking in strength (Fig. 24). In practically every instance muscle exercises or correcting prisms may then be prescribed with definite knowledge of requirements, as further described in the following paragraphs.

A binocular muscle test made with the phorometer, Maddox rod and distant muscle-testing point of light might quickly indicate six degrees of exophoria, both before and after the optical correction is made. While this would doubtless be the correct amount of the manifest imbalance, it would be a difficult matter to ascertain which muscles caused the disturbance. To determine this important question, the monocular or duction test should be invariably employed.

DIAGNOSING A SPECIFIC MUSCLE CASE

Assuming, for example, a specific case where six degrees of exophoria was determined in the binocular test that the muscle findings in the duction test show right adduction of twenty-four degrees, with an accompanying abduction of eight degrees; likewise a superduction and subduction of two degrees for each eye. With the aid of a chart or diagram—which should be made in every case—a comparison of these figures would indicate an exophoria of approximately six degrees, with a corresponding weak left internus (Fig. 24). This not only shows the muscle pull of each eye individually, but a comparison of the two eyes as indicated by the dotted lines. Thus the relationship of the two eyes, and their corresponding muscles is quickly indicated.

Fig. 24—Duction chart should be made in every case. Above readily shows existence of muscular imbalance and proves subduction and superduction for both eyes are equal; otherwise hyperphoria would be disclosed. Also note abduction for both right and left eye are equal, otherwise esophoria would be disclosed. Also note adduction for right eye is 24° while left is but 18°, proving a case of 6° of exophoria with a left weak internus.

A glance at the above diagram discloses the following three important facts, all of which should be known to the refractionist before a single thought can be devoted to the correcting of the case:

1. 6° exophoria is the amount of the insufficiency.

2. 18° adduction (which should be 24°).

3. Left weak internus.

As previously stated, the power to converge is normally rated 3 to 1, or 8 to 24, as shown above, while the power of the eye to look upward, is equal to the power to look downward. The diagram accordingly proves that the muscles of the right eye are in perfect balance, having equal muscular energy.

A comparison of the left eye shows adduction of 18 degrees with an abduction of 8 degrees, proving a lateral insufficiency because the ratio is less than 3 to 1; and the muscles of the left eye are at fault. The power of 2 degrees superduction and 2 degrees subduction, proves that no weakness exists in the vertical muscles.

After making the duction test for each eye individually, a comparison of both eyes in relationship to each other may be more readily determined by following the dotted lines (Fig. 24).

As previously stated, it is the inability of the two eyes to work together that causes the imbalance, so that if both eyes were normal, the adduction, abduction, superduction and subduction of the two eyes would agree.

The duction chart (Fig. 24.) also shows that the corresponding muscles of each eye agree—with the exception of the adduction of the right eye and the left eye. This proves that the left internus is weak, measuring only 18 degrees instead of 24 degrees; it further proves the 6 degrees of exophoria in the monocular test, as was quickly and more readily determined in the binocular test.

Likewise, in cases of esophoria, hyperphoria, or cataphoria, the making of definite muscle measurements independently through the prescribed method would show through the merest glance at a similar diagram which muscle or muscles were relatively out of balance. Heterphoria of almost any type, or tendencies other than normal, may be fully investigated by making a thorough and separate test of each muscle.

Where an imbalance exists, a rapid test may be employed to distinguish a pseudo or false condition from a true condition. This is accomplished by first placing the two Maddox rods (both the red and white) so that the rods lie in a vertical position. If the two lines fuse, we have determined the existence of a false condition caused by a possible error of refraction or nerve strain. If the lines separate, we have determined a true muscular condition, and then only should the second method of muscular treatment follow.

SKI-OPTOMETER MODEL 205

Same as Model 215 but Automatic Cylinder Arrangement omitted.

Embodying Spherical Lenses Combined with Appliances for Testing and Correcting Muscular Imbalance.

CHAPTER IX

FIRST METHOD OF TREATMENT

OPTICAL CORRECTION

THE mere determination of the degree of an imbalance, or even the diagnosis of a patient’s trouble, is not sufficient. If relief is to be secured, something more must be accomplished.

As previously stated, muscular imbalance may be corrected through one of the four following rules or methods, each explained in their relative order:

- 1—Optical Correction

- 2—Muscular Exercise

- 3—Use of Prism Lenses

- 4—Operative Methods

ESOPHORIA

To correct a case of muscular imbalance, where six degrees of esophoria has been determined, the first rule of making the test for optical correction with the Ski-optometer’s spherical and cylindrical lenses, would be in the line of routine. The binocular test made with the phorometer and combined use of the red Maddox rod would have determined the six degrees of esophoria.

The reason for making the binocular muscle test before and after the optical correction is because an imbalance is often aggravated or benefited by the correcting lenses. The optical correction frequently eliminates the need for further muscular treatment.

For example, we will assume that the optical correction tends to decrease the degree of esophoria from six degrees to four degrees. According to the previously mentioned rule for correcting cases exceeding one degree in hyperphoria, three degrees in exophoria and five degrees in esophoria, the condition would indicate that of being “left alone.” Just what is taking place should be fully understood—its cause as well as its effect.

Fig. 25—Comparative diagram showing how a decentered lens before a centered eye has the same effect as a centered lens before a decentered eye.

When not otherwise specified, accurately centered lenses are of primary importance. The pupil of the eye should be directly behind the center of each lens (Fig. 25).

Figure “A” of the latter sketch illustrates a perfectly centered lens—its center indicated by a cross, the circle representing an eye directly behind it. Figure “B” illustrates a perfectly centered pupil behind a prism, with its center designated by a cross. To ascertain how the centered spherical lens takes the place of a prism, Figure “C” should be compared with Figure “B”; this will show that the eye is decentered, while the lens is centered. A further comparison will prove that the results in Figures “B” and “C” are identical, the correcting lenses having practically the same effect through the decentration of the eye as if a prism were prescribed, nature supplying its own decentration.

TREATMENT FOR CORRECTING ESOPHORIA

IN CHILDREN

In case of esophoria, regardless of amount, slightly increased spherical power is frequently prescribed for children. This will naturally blur or fog the patient’s vision, but in their effort to overcome the blur, accommodation is relaxed, usually tending to correct the muscular defect.

In such cases, as a rule, a quarter diopter increased spherical strength may frequently be added for each degree of esophoria as determined before the optical correction was made. In a case of 6 degrees of esophoria, the refractionist may prescribe +1.50 diopter spherical added to the optical correction, which, let us assume, is +1.00 sph. = -1.00 cyl. ax. 180°, so that the treatment glasses would be +2.50 sph. = -1.00 ax. 180° (See Procedure on Page 74).

At the end of each three months’ period, the patient should be requested to return, when the binocular and the duction test should again be made, comparing results with the work previously accomplished. An improvement tending to build up the left weak externus will possibly permit of a decrease of the excessive spherical power, so that excessive spherical power is reduced until completely removed, in all probability overcoming the muscular defect. Esophoria is almost invariably a false condition and frequently is outgrown under this treatment as the child advances in years. On the other hand, esophoria uncared for in the child may tend to produce exophoria in the adult.

HOW OPTICAL CORRECTION TENDS TO

DECREASE 6° ESOPHORIA IN A CHILD

Assume binocular muscle test made before optical correction shows

6° Esophoria.

+1. Sph. = -1. Cyl. Ax. 180.

Next, locate faulty muscle by making a duction test, which shows how abduction of left eye is made to equal that of right eye, change being made quarterly with treatment lenses in accordance with following rule. Note as abduction is increased, esophoria is reduced.

Rule—prescribe a quarter diopter increased sphere for each degree of imbalance or 0.25 × 6 equals:

+1.50 added to optical correction.

1/1/19 (assumed date) prescribed treatment lenses equal:

+2.50 = -1. × 180°.

4/1/19 (3 months later) assuming abduction has increased from 2° to 3° showing difference of 5 Es. or 0.25 × 5. equals +1.25 added to optical correction, prescribed treatment lenses equal:

+2.25 = -1. × 180.

7/1/19 (3 months later), assuming abduction has increased from 3° to 4° showing difference of 4° Es. or 0.25 × 4 equals +1.00 which added to optical correction would make prescribed treatment lenses equal:

+2.00 = -1. × 180.

And so on, every three months treatment lenses are prescribed until both right and left eye show 8° of abduction. In this way the treatment lenses are reduced to original correction of +1.00 = -100 × 180. This would have required six changes of lenses, three months apart—thus consuming 18 months time.

CHAPTER X

SECOND METHOD OF TREATMENT—

MUSCULAR EXERCISE

MADE WITH TWO ROTARY PRISMS

AND RED MADDOX ROD

EXOPHORIA

IF A case is one of exophoria of six degrees, where the second method of treatment or muscular exercise is in line of routine, it is essential to first determine through a duction test and the preparation of the diagram exactly which one of the four muscles are faulty (Fig. 24).

Having determined, with the aid of the diagram, first, the existence of 6 degrees of exophoria; second, 18 degrees of adduction; third, a weak left internus—the next procedure is to determine what degree of prism will enable the patient to obtain single binocular vision, with both eyes looking “straight.”

To determine this, place both of the Ski-optometer’s rotary prisms in position with the handle of each pointing outward horizontally. The red line or indicator of each prism should then be placed at 30° of the outer scale (Fig. 26).

The red Maddox rod should be horizontally positioned before the eye, the white line on indicator pointing to 180° of the scale (Fig. 27). The strength of the rotary prism before the right eye should thereupon be reduced by rotating the prism indicator or red line toward the upper zero (0) to a point where the patient first sees the red streak—assuming that the red line appears at 42 degrees, that is 30 degrees before the left eye and 12 degrees before the right.

Fig. 26 (A and B)—First position of rotary prisms to determine amount of prism exercise to be employed for building up the weak muscle.

The prism should then be still further reduced until the vertical streak produced by the Maddox rod directly bisects the muscle testing spot of light. Assuming that this point be thirty-eight degrees, which is four degrees less, single binocular vision is produced.

Fig. 27—Position of red Maddox rod used in conjunction with Fig. 26 for prism exercising.

For example, sixty degrees of prism power (the combined power of the two rotary prisms) will usually cause complete distortion. Therefore, as outlined in Figure 28, the patient, seeing only out of the right eye, will detect nothing but a white light. By gradually reducing the strength of the prism before the right, which is the good eye, the patient will eventually see a red streak off to the left. A continued and gradual reduction to a point where the red streak bisects the white light, will determine how much prism power is required for the patient to obtain single binocular vision, thus establishing the same image at the same time on each fovea or retina (Fig. 20).

This has taught the patient to do that which he has never before accomplished. Therefore, after having been taught how to make the two eyes work in relation to each other, the natural tendency thereafter will be to strive for the same relationship of vision with both eyes. The refractionist should then aim to reduce the excessive amount of prism required to give binocular vision, which can be accomplished by muscular exercise.

It must always be remembered before the refractionist is ready to employ the muscular exercise or second method, that the degree of prism required to give the patient single binocular vision must be determined with the optical correction in place. The exercise must be practised daily in routine, a daily record being essential.

AN ASSUMED CASE

We will assume a case where 42 degrees is required to enable the patient to first see the red streak as produced by the Maddox rod to the extreme left. Through a continued gradual reduction of 4 degrees (or to 38 degrees), we next learn that the streak was carried over until it bisected the white spot of light, giving single binocular vision and producing a position of rest.

Fig. 28—Simplified chart showing the prism action employed in developing a weak ocular muscle through alternating prism exercise. Either side of 38° in excess of 4° causing diplopia.

The patient has now established the limitation of the exercise, which is four degrees, this limitation being determined by the difference between the point where the streak was first seen to the extreme side and where it bisected the spot. The same amount of four degrees should then be used for the opposite side, thus reducing the prism strength to 34 degrees.

This again produces diplopia, because of the lesser amount of prism power employed to give single binocular vision. The refractionist should then return to 38 degrees, where single binocular vision had originally been determined (Fig. 28), alternating back to 42, returning to 38, over to 34, back to 38, and so on. This procedure should be employed once a day just after meals for about five minutes, and repeated ten times, constantly striving for a slight reduction of prism power from day to day.

EFFECT OF MUSCULAR EXERCISE

This muscular treatment, or constructive exercising, should enable the patient to overcome his amount of four degrees in either direction in about a week. Hence in the case showing 38 degrees for single binocular vision, results may be looked for in about nine weeks—four degrees divided into 38 degrees. While the patient is undergoing the treatment, which is nothing more than the strengthening of the interni muscles or developing adduction, it is natural to believe that the amount of imbalance is likewise being conquered. This, however, is readily determined from time to time by making the binocular muscle test with the phorometer and Maddox rod, as well as the duction chart test (Fig. 24), as previously outlined.

To fully appreciate the effect of this muscular treatment, the reader need only hold his head in a stationary position, casting his eyes several times from the extreme right to the extreme left, not failing to note the apparent muscular strain. On the other hand, with the aid of the Ski-optometer’s rotating prisms, the refractionist not only has complete control of the patient’s muscles at all times, but scientifically accomplishes muscular exercise without any tiresome strain, overcoming all possible exertion.

After the case in question has been reduced to 30 degrees, having no further use for the rotary prism, it may be removed from before the right eye and the same exercising procedure continued as before with the remaining left side rotary prism by reducing its power, until it is likewise down to zero.

Having reduced both prisms to zero, each prism should again be placed in position with zero graduations vertical and the prism indicator on upper zero. Both prisms should then be turned simultaneously about four degrees toward the nasal side of the patient, thus tending to jointly force corresponding muscles of both eyes.

HOME TREATMENT FOR MUSCULAR EXERCISE—

SQUARE PRISM SET USED IN CONJUNCTION

WITH THE SKI-OPTOMETER

Where a patient is unable to call each day for this muscular treatment or exercise, the work will be greatly facilitated by employing a specially designed set of square prisms ranging in strength from ½ to 20 degrees for home treatment. As in the case previously cited, it is necessary to carefully instruct the patient that the interni muscles must be developed, hence prism base out with apex in must be employed. Attention should then be directed to a candle light, serving as a muscle testing spot of light and stationed in a semi-dark room at an approximate distance of twenty feet.

Having determined through the Ski-optometer the strength of the prism required after each office treatment, its equivalent should then be placed in a special square prism trial-frame which permits rotation of the prism, although the patient is frequently taught to twirl the lens before the eye. This exercise may be continued for about five minutes each day.

The patient should also be instructed to call at the end of each week, when the work may be checked by means of the Ski-optometer’s rotary prisms, making the duction test as previously explained and outlined in Fig. 24. It is then possible to determine whether or not satisfactory results are being obtained. Otherwise the exercise should be abandoned.

Should the second method employed in the work of muscular imbalance not prove effective, the third method requiring the use of prisms would be next in routine.

CHAPTER XI

THIRD METHOD OF TREATMENT—PRISM LENSES

WHEN AND HOW EMPLOYED

AS stated in the preceding chapter, on ascertaining the failure of the second muscular treatment or method, prisms are employed for constant wear. When prism lenses are used, whether the case is exophoria or esophoria, or right or left hyperphoria, it is always safe to prescribe one-quarter degree prism for each degree of prism imbalance for each eye. For example, in a case of 6 degrees of esophoria, a prism of 1½ degree base out should be prescribed for each eye; or in 6 degrees of exophoria, employ the same amount of prism, but base in. In right hyperphoria, place the prism base down before the right eye and up before the left, and vice versa for left hyperphoria.

It is not always advisable, however, to allow the patient to wear the same degree of prism for any length of time. Many authorities suggest a constant change with the idea that a prism is nothing more than a crutch. Should the same degree be constantly worn, even though it afforded temporary relief, the eye would become accustomed to it and the purpose of the prism entirely lost. Prisms should be prescribed with extreme care, their use being identical with that of dumb-bells, where weight is first increased to maximum and subsequently reduced, viz.:

PRISM REDUCTION METHOD

Where prisms are prescribed, it is considered good practice to make a binocular muscle test and the duction test (Fig. 24) at the end of each three months’ period, employing the phorometer, Maddox rod, and rotary prisms, as already explained.

If the condition shows any decrease, the prism degree should be proportionately decreased. For example, in the case originally showing 6 degrees of exophoria, one-quarter degree prism for each degree of imbalance was prescribed, or 1½ degree for each eye. If the same case subsequently indicated 4 degrees, only one degree for each eye should be prescribed—and so on, a gradual reduction of prism value being constantly sought.

Except in rare cases, prisms should not be prescribed with the base or apex at oblique angles, as the eye is rarely at rest with such a correction. An imbalance may be caused by a false condition in one rectus and a true imbalance in the other, giving one the impression that cyclophoria exists, as explained in a following chapter.

Having now employed the three methods, the refractionist can readily understand that a marked percentage of muscular imbalance cases may be directly benefited through the aid of the Ski-optometer. If these three methods of procedure fail, there is nothing left but the fourth and last method—that of operative procedure.

CHAPTER XII

A CONDENSATION OF PREVIOUS CHAPTERS

ON THE PROCEDURE FOR MUSCLE TESTING

WITH THE SKI-OPTOMETER

THE present chapter, intended for those desiring a synopsis or condensed summary of muscular imbalance work, should prove of the utmost assistance to the busy refractionist. Muscular imbalance work can be successfully conducted if the following routine is studied and memorized, with the Ski-optometer constantly before the reader. The chapters containing the corresponding figures and diagrams or illustrations will then be readily comprehended. It is also important to carefully note the captions under each diagram.

1. Without any testing lenses before patient’s eyes, direct attention to a 20-foot distant muscle testing spot of light (Fig. 9).

2. Place phorometer handle vertically (Fig. 16).

Place red Maddox rod vertically (Fig. 15). Patient should see a white spot of light, and a red horizontal streak (Fig. 17).

Simply turn phorometer handle until horizontal streak bisects white spot of light. Pointer then indicates amount of deviation on red scale. Ignore cases less than 1° hyperphoria, whether right or left designated by (R. H.—L. H.).

3. Place phorometer handle horizontally (Fig. 19).

Place red Maddox rod horizontally (Fig. 18). Patient should see a white spot of light and a vertical red streak (Fig. 20).

Simply turn phorometer handle until red streak bisects spot of light. Pointer indicates amount of deviation on white scale, whether esophoria or exophoria designated by (Es—Ex).

4. Ignore all exophoria cases, less than 3°.

Ignore all esophoria cases, less than 5°—except in children, ignore less than 3° of esophoria.

5. Always make the above or binocular muscle test—with phorometer and red Maddox before optical correction or (test for spheres and cylinders) and again after optical correction where case shows more than 1-3-5 rule, to determine whether muscles are aggravated or benefited.

6. In cases showing more than the 1-3-5 rule, shown in above No. 4, make monocular duction test first with rotary prism before patient’s right eye,—then with rotary prism before left eye to find faulty muscle and determine which eye is affected.

7. To test adduction, prism base out is required. Rotary prism’s red line or indicator should be rotated from zero outwardly. To test abduction, base in is required. Indicator should be rotated inwardly from zero (Fig. 22). Power of adduction as compared with abduction, is normally 3 to 1—usually rated 24 to 8.

8. To test superduction, base down is required. Rotary prism’s line or indicator should be rotated downward from zero. To test subduction, base up is required. Indicator should be rotated upward from zero. Power of superduction as compared with subduction, is normally equal—usually rated 2 for each (Fig. 23).

9. Direct patient’s attention to largest letter on distant chart, usually letter “E,” rotating red line indicator of rotary prism outlined in above No. 7 and No. 8, until diplopia is first procured.

10. The use of a duction chart on a record card, quickly designates pull for each of four muscles (Fig. 24), illustrating an assumed case of—

- 1st—6D of Exophoria.

- 2nd—18° adduction (which must be developed to 24°).

- 3rd—Patient has a left weak internus.

11. Employ First Method—Optical Correction—to effect treatment.

12. Assuming a case of a child with 6° of esophoria—8° of right abduction and 2° left abduction indicating a left weak externus, prescribe a quarter diopter increased plus spherical power for each degree of imbalance, thus adding +1.50D spherical to optical correction. This is the first method of treatment. This requires a thorough reading of Chapter IX on Treatment for Correcting Esophoria in Children and a careful study of the formula. For synopsis see Page 74.

FOUR METHODS OF TREATING AN IMBALANCE CASE

WHEN THE PRECEDING ONE FAILS

1st—Optical correction;

2nd—Muscular exercise or treatment;

75% ARE CURABLE WITH FIRST AND SECOND METHODS.

3rd—Prisms;

5% ARE CURABLE WITH THIRD METHOD.

4th—Operation;

20% ARE CURABLE WITH FOURTH METHOD.

13. When first method of treatment fails, EMPLOY SECOND METHOD—Muscular Exercise—to effect treatment.

1st—Find degree of prism patient will accept to produce single binocular vision with optical correction on, placing both rotary prisms in position, handles horizontal, red line on 30° of temporal scale of each, giving total value to 60° (Fig. 26a and b).

2nd—Also place red Maddox rod before patient’s eye (rods horizontal) (Fig. 18), calling patient’s attention to usual muscle testing spot of light.

3rd—Reduce prism before good eye until red streak appears, noting degree (which we assume shows 42° the combined total value of both prisms) slowly continue to decrease prism until streak bisects spot. Assume this shows total of 38°. Either side of 38° in excess of 4° (38 to 42) produces diplopia. Prisms must only be rotated from 38° to 42° back to 38° over to 34°—back to 38° over to 42°—back again to 38° and so on—exercise to be continued daily ten times for five minutes (Fig. 28).

4th—At end of each week, duction test should again be made. Duction chart should show a tendency to reduce exophoria by a gradual building up of adduction, approximately one week is usually sufficient to teach patient to hold streak within the spot (between 38° and 42°). Exercise to be continued until both prisms are worked down to zero. Exercise tends to teach patient how to establish same image on each fovea or retina at same time.

5th—If patient is unable to call daily for treatment, employ home treatment. (Read “Home Treatment for Muscular Exercising,” Page 82).

Employ THIRD METHOD—Use of Prisms for Constant Wear to effect treatment.

PRISMS

1st. Where a case cannot be reduced through use of first two methods, as for example in a case of 6° of exophoria, prescribe ¼ of amount of imbalance (¼ × 6 = 1½°) for each eye—base in—or esophoria base out, hyperphoria base up on eye affected.

2nd. Advise patient to call every three months and make duction test (Fig. 24). If no improvement in condition, after wearing prisms six months, operative means is suggested.

Assume a case is benefited, reduce prism power according to rule; ¼D prism for each degree of imbalance.

CYCLOPHORIA

This work being of a technical nature, it is deemed best for the reader to study Chapter XIII and XIV.

CHAPTER XIII

CYCLOPHORIA

MADE WITH MADDOX RODS

AND ROTARY PRISMS

CYCLOPHORIA, a condition affecting the oblique muscles of the eye, is caused by its rotation. It is detected in the following manner by the combined use of the red and white Maddox rods and the rotary prism.

Fig. 29—Position of rotary prism for producing diplopia in testing cyclophoria with prism placed at 8° base up.

Darken the room and direct the patient’s attention to the usual muscle-testing spot of light, located approximately twenty feet away and on a direct plane with the patient’s eye. The optical correction, if one is required, should always be left in place—just as in making other previously described muscle tests.

The rotary prism should then be brought before the patient’s right eye with the handle-pointing upward and with zero graduations horizontal. The indicator or red line should then be rotated upward from zero to eight upon the prism scale, creating the equivalent of a prism of 8 diopters with base up (Fig. 29). This normally caused diplopia, although in some cases it may be necessary to place the prism at 10 or 12 degrees before diplopia is produced.

Fig. 30—(A. and B.)—First position of both Maddox rods used in conjunction with Fig. 29 for determining cyclophoria.

The red Maddox rod should then be brought into operative position before the patient’s left eye (Fig. 30a) and the white Maddox rod before the patient’s right eye, (Fig. 30b) setting each one so that the rods lie in a vertical position with their white line on the large red zero (0).

The patient should now see two separate and distinct streaks of light, one appearing below the other.

DETERMINING CYCLOPHORIA

Fig. 31

Fig. 34

Fig. 32

Fig. 35

Fig. 33

Fig. 36

Figs. 31-36—Diagram showing how streaks appear to patient, as produced by the Maddox rods in testing for cyclophoria.

Should there be no cyclophoria of the right eye, the streaks will appear in a horizontal plane parallel to each other (Fig. 31).

Should the red streak appear horizontally to the left eye, and the white streak seen by the right eye appear at an angle therewith, cyclophoria of the right eye would be indicated (Fig. 32).

In brief, should the white streak dip towards the patient’s left side, the case would be one of right plus cyclophoria (Fig. 32); whereas right minus cyclophoria would be indicated should the white streak dip to the patient’s right side (Fig. 33).

Next, setting the rotary prism of 8 degrees, placed base up before the patient’s left eye, the red streak should appear below the white one. Should the two streaks appear horizontally, parallel with each other, there would be no cyclophoria of the left eye (Fig. 34).

If, however, the upper or white streak should appear horizontal, and the lower or red streak at an angle therewith dipping toward the patient’s right side, the left eye would be cyclophoric and the case would be one of left plus cyclophoria, as the chart indicates (Fig. 35).

Should the red streak dip in toward the patient’s left side, left minus cyclophoria would be designated (Fig. 36).

The patient would instinctively describe, with pointed finger and hand motion, the position of the “dipping” line just as one would describe a spiral staircase. Should this test determine that no cyclophoria exists in either eye, there would be no necessity for further tests.

Some authorities claim that both Maddox rods should be of the same color, so as to more readily assist the patient to fuse the two objects. If the reader so desires, he can readily place a red lens from the trial-case in the forward cell of the instrument.

The characters plus and minus in cyclophoria merely refer to plus as signifying a tendency toward the temporal side; minus indicating a tendency toward the nasal side. This has no bearing on “convex” and “concave,” which are frequently designated as “plus” and “minus.”

The test for cyclophoria is particularly essential, proving of utmost importance where the patient requires an astigmatic correction with the cylinder axis in oblique meridian. The case should then be investigated in every instance by making a thorough and separate test of each eye for cyclophoria.

In a case where cyclophoria is determined, the trouble may be caused by the functioning of other muscles, through the drain of nerve force, thus disturbing the harmony of every muscle action.

Cyclophoria is frequently caused by an imbalance of two recti, giving an oblique pull. In most cases, it is merely necessary to release the torsion, as described in the following chapter.

CHAPTER XIV

CYCLODUCTION TEST

MADE WITH THE COMBINED USE OF

THE TWO MADDOX RODS

HAVING determined that cyclophoria exists, as previously outlined, the next step would be to make a cycloduction test, or a test of the oblique muscles individually. Maddox rods, both red and white, should be placed in position with the rods horizontal—the plus and minus sign at 90 degrees on the scale (Fig. 37). The patient’s attention should be directed to the usual muscle-testing spot of light, when a vertical band of light will appear to the patient, as shown in Fig. 38.

Fig. 37—(A. and B.)—Primary position of combined use of both Maddox rods for determining cycloduction test.

Fig. 38

Fig. 39

Fig. 40

Figs. 38-40—Diagram showing position of streaks produced by Maddox rods as they appear to patients in making cycloduction tests.

To measure the duction range of the inferior oblique of the right eye, it is merely necessary to slowly rotate the Maddox rod before the right eye upward at its nasal end to the point where the band of light breaks so as to resemble a letter “X”. This gives in degrees the amount of right plus cycloduction, as indicated on the temporal scale, when it will appear to the patient, as shown in Fig. 39.

The Maddox rod should then be restored to its original position, with the plus and minus on the 90 degree line of the scale (Fig. 37), and rotated upward at the temporal end until it again takes the form of the letter “X”. (Fig. 40.) The position of the indicator will now denote the amount of right minus cycloduction, or duction range of the right superior oblique muscles. Having determined the duction range of the oblique muscles of the patient’s right eye, both Maddox rods should be placed in original position with rods horizontal and plus and minus sign on 90° of scale, as shown in Fig. 37.

The Maddox rod before the left eye is then employed exactly in the same manner as before when the test for the right eye was made. A plus cycloduction of the left eye would be indicated, as shown in Fig. 40, while a minus cycloduction of the left eye would appear to the patient, as shown in Fig. 39.

By recording a comparison of each eye, as explained, it will be found that the range of duction usually averages five to twenty degrees on either side of the 90 degree line, as indicated on the scale surrounding the Maddox rods.

It will be recalled that cyclophoria was only to be looked for in oblique astigmatic cases. It is frequently possible to correct the patient’s trouble, by changing the axis of the cylinder, before one or both eyes, a minus cycloduction signifying a change of axis towards 180° while a plus toward 90°, according to the amount lacking in full duction power. It is also well to exercise the oblique muscles through a rotation of the Maddox rod before the affected eye, whether it be one or both that is lacking in full duction power, until the required amount is reached to equal its fellow member.

For a more exhaustive treatise the author suggests a reading of Dr. Savage’s work on the subject.

TREATMENT FOR CYCLOPHORIA

As previously stated, it often proves of great benefit to employ a muscular exercise where a patient has an existing cyclophoria of either one or both eyes, results derivable through the exercise of the recti muscles having been previously detailed.

To exercise the oblique muscles of the right eye, both Maddox rods should be placed in the original position employed for making cycloduction test, as previously explained (Fig. 37). This causes the patient to see but one band or vertical streak (Fig. 38).

The Maddox rod, placed before the right eye, should be slowly rotated inward from ninety degrees to a point on the scale where the single streak of light breaks, when it should again be returned to ninety degrees. This causes a contraction and relaxation of the muscles in the form of an exercise and should be repeated ten times—about five minutes each day. By employing the Maddox rod before the left eye in precisely the same manner, its oblique muscles will be exercised.

To determine whether or not this form of exercise is beneficial to the patient, the weekly cycloduction test, as previously described, should be made and compared with the original findings.

SKI-OPTOMETER MODEL 235

A Compact Phorometer and Trial-Frame.

For Testing and Correcting Muscular Imbalance—

Providing a Comfortable Form of Trial-Frame.

CHAPTER XV

MOVEMENTS OF THE EYEBALLS

AND THEIR ANOMALIES

AFTER a careful study of the foregoing chapters, the refractionist may desire further knowledge concerning muscular imbalance—a matter in which the Ski-optometer plays an exceptionally important part.

It should be remembered that it is only the general utility of the instrument, plus one’s knowledge of refraction and individual diagnosis that enables the refractionist to attain maximum efficiency in every examination, a fact which largely accounts for the following chapter.

MONOCULAR FIXATION

When we view an object directly, so that it appears to be more distinct than surrounding objects, we are said to “fix” or “fixate” it.

As the fovea is normally the most sensitive part of the retina, affording by far the most distinct vision, “fixation,” in the great majority of cases, is so performed that the image of the object that is “fixated” falls upon the fovea of the eye that is “fixing.” This is known as central or muscular “fixation.”

When central vision is absent, however, the patient is compelled to see with a portion of the retina outside of the fovea. The eye must then be so directed as to cause the image of the object to fall on this outlying portion of the retina. This is termed “eccentric fixation,” and usually denotes that vision is exceptionally poor.

The ability to “fix” is apparently acquired in early infancy by constant practice in looking at objects. Any marked interference with vision, particularly with central vision—present at birth or soon thereafter—will tend to prevent the acquisition of this ability, and in extreme cases the eye does not learn to “fix” at all, but aimlessly wanders in all directions.

BINOCULAR FIXATION

We habitually use the eyes together, fixating with both at once; that is, we direct the eyes in such a way that the image of the object to which the attention is directed falls on the fovea of each eye.

Where both eyes are accurately directed to an object at which one or both are looking, the condition is known as “binocular fixation,” which is commonly understood to mean that both eyes are straight.

The ability to produce and maintain binocular fixation—to keep both eyes directly straight—is acquired early in life. The impulse to maintain it grows with exercise, and soon becomes so strong that after the age of infancy binocular fixation is present in the great majority of persons, and in most of them is present all the time.

Binocular fixation must be distinguished through three conditions—orthophoria, heterophoria and squint.

ORTHOPHORIA

This is the condition in which both eyes look straight at the same object, whether both see it or not. There is not the slightest tendency of deviation.

HETEROPHORIA

This is the condition in which both eyes keep looking straight at the same object so long as both see it; but as soon as one eye is excluded from vision (as by a screen) that eye deviates. This is then a tendency of deviation which is strong enough to become manifest when either eye is covered, but which is abolished or overcome by the compelling impulse of binocular fixation as soon as both eyes are used for seeing. A heterophoria thus produces a maximum deviation. The deviation is also said to be latent, since it is absent under ordinary conditions and is brought to light only under special conditions. A common though improper term for heterophoria is “insufficiency.”

SQUINT

Squint is the condition in which there is so great a tendency to deviation that even when both eyes are uncovered, one deviates and only one “fixes.” It differs, therefore, from heterophoria in that the deviation it produces is obvious under ordinary conditions.

Squint is also called strabismus, or heterotropia. In other words, in orthophoria there is binocular fixation all the time and under all conditions; in heterophoria it is present only when the two eyes are uncovered, so that both see the object looked at; while in squint it is not present at all.

Or, in still plainer terms, in orthophoria both eyes are straight all the time; in heterophoria both are straight, but only so long as both are uncovered; and in squint only one eye is straight, no matter whether both eyes are uncovered or not.

In squint, while binocular fixation is altogether absent, the ability to perform monocular fixation is almost always preserved; i.e., the squinting eye will “fix” at once if the other eye is covered. It is only when there is marked amblyopia, particularly as the result of a central scotoma (or spot on the cornea in the line of vision) that the squinting eye loses its power to fix at all, and wanders uncertainly about, receiving impressions now on one, now on another portion of the retina.

The term imbalance is often used to denote the two conditions opposed to orthophoria; i.e., to denote collectively heterophoria and squint.

VARIETIES OF HETEROPHORIA AND SQUINT

1. Classification According to Direction of Deviating Eye: Heterophoria and squint may be classified according to the direction assumed by the deviating eye. Thus we have the following varieties of heterophoria:

HETEROPHORIA

LATERAL DEVIATIONS

Either eye deviates

In, or toward the nose

Esophoria

Out, or toward the temple

Exophoria

VERTICAL DEVIATIONS

The right eye goes up and the left down

Right Hyperphoria

The left eye goes up and the right down

Left Hyperphoria

In rare cases of vertical heterophoria, each eye has either an upward tendency (anophoria) or a downward tendency (cataphoria). These cases must not be confused with anatropia and catatropia. In anaphoria and cataphoria, there is binocular fixation when both eyes are uncovered, while in anatropia and catatropia one of the eyes squints. This shows the following squint condition:

SQUINT

LATERAL SQUINT

The deviating eye turns in, or toward the nose:

Esotropia (Strabismus convergens—Convergent Squint)

The deviating eye turns out or toward the temple:

Exotropia (Strabismus divergens—Divergent Squint)

VERTICAL SQUINT

The deviating eye turns up:

Hypertropia (Strabismus sursumvergens) (Right or left)

The deviating eye turns down:

Hypotropia (Strabismus deorsumvergens) (Right or left)

In addition to these lateral and vertical deviations, conditions exist in which the vertical meridian of one eye, instead of maintaining its parallelism with the vertical meridian of the other, either forms (or tends to form) an angle with it (cyclotropia), but is kept in position through muscular effort (cyclophoria.)

Cyclotropia is usually due to paralysis of one of the ocular muscles, causing the vertical meridian of the affected eye to be tilted out or toward the temple (extorsion) or in toward the nose (intorsion). A tilting of the vertical meridian toward the right is also called dextrotorsion (or positive declination); and to the left, levotorsion or negative declination.

2. Constant, Intermittent and Periodic Deviations: A deviation, whether squint or heterophoria, may be present at all times (constant), or occasionally present and occasionally absent (intermittent). In this case we may have heterophoria alternating with orthophoria, or heterophoria alternating with squint; or squint alternating with orthophoria. We also find variations such as a squint for near and a heterophoria or orthophoria for distance; or a heterophoria for near and orthophoria for distance; or a constant squint for near and an intermittent squint for distance, etc. Again, a deviation may be periodic, in that its amount for distance may greatly exceed that for near, or vice versa.

Opposed to a periodic deviation is one which is present, and in about equal amount, both for distance and near. Such a deviation, whether squint or heterophoria, is called “continuous.”

3. Alternating and Uniocular Squint: An alternating squint is one in which when both eyes are uncovered, so that both have a chance to “fix”; sometimes the right eye will deviate, sometimes the left. In uniocular (less properly monocular) squint, under the same conditions, one eye, either the right or the left, always “fixes” and the other always deviates. A uniocular squint is denoted as right or left, according to whether it is the right or left eye which deviates.

A TYPICAL REFRACTION ROOM—

THE WOOLF SANITARY ALL-METAL EQUIPMENT

Installation comprising: Ophthalmic Chair, complete with Ski-optometer, Test Letter Cabinet, Asceptic Trial-Case Cabinet, Muscle Testing and Skioscopic Lamp, Ophthalmometer, Perimeter, Adjustable Tables, Adjustable Stool.

4. Comitant and Non-Comitant Deviations: In some varieties of heterophoria and squint, the amount of deviation is the same in all directions of the gaze, so that the angle between the visual line of one eye and that of the other remains the same, no matter which way the eyes are turned. Such deviations are called comitant or non-comitant, because one eye accompanies and keeps pace with the other in all its movements. In other cases, the deviation changes as the eyes are moved in different directions, so that the angle between the two visual lines constantly varies. Such deviations are termed concomitant. Usually in a non-comitant squint the angle of deviation increases in a regular way as the eyes are moved in one direction and decreases as they move in the direction opposite.

In cases of long standing, however, the squinting eye, particularly when very amblyopic, wanders in an uncertain way and apparently quite without reference to the movements of the other eye.

CHAPTER XVI

LAW OF PROJECTION

THE movements of the eye are designed primarily to effect fixation—that is, to bring upon the macula the image of the object that we wish to look at. When this has been accomplished, we know as a result of long experience, the direction of the object looked at and also direction of other neighboring objects. This knowledge is doubtless afforded us, in part, by our muscle sense. Thus we know that an object, A, is straight in front of us because we can see it sharply without moving either the head or the eyes from the position of rest or equilibrium; and we know that an object, B, is on the right of us because to see it sharply we have to move either the head or the eyes to the right, thus altering the muscular condition from one of rest to one of tension. But without moving either head or eye, we also know, while still looking at A, that B is to the right, for the image of B is then formed on a portion of the retina situated to the left of the macula. From long experience we also know that an image so situated means an object placed on our right. Moreover, the farther to the left of the macula the image B is, the farther to the right do we judge B itself to be.

Similarly, if B is so placed that its image falls below the macula, we then know B itself is really above A, which forms its image on the macula; and if the image of B is above the macula, we know that B itself is below A.

The table on page 116 is suggested as a guide in cases of muscular imbalance:

SUPPRESSION OF IMAGE

All deviations should be and probably are primarily associated with diplopia. Yet in the great majority of cases of established squint, especially convergent squint, there is no double vision. This is due to the mental suppression of the image by the squinting eye. In such cases all attempts to evoke diplopia by our tests may be futile, the patient not appreciating the presence of double images even when they are widely separated by prisms. Moreover, this suppression usually persists after the squint is cured, so that even though there are two retinal images of the same object, the mind perceives but one and ignores the other, just as though it were not present. In this case there is no true stereoscopic, or solid, vision.

MONOCULAR DIPLOPIA

Binocular diplopia, due to deviation of the eyes or to prisms, must be distinguished from monocular diplopia, or the condition in which the patient sees double with one eye alone. This occurs as the result of astigmatism, plus spherical aberration and other conditions found occasionally in squint. It can readily be differentiated by the fact that binocular diplopia disappears when the patient shuts either eye; while monocular diplopia, of course, does not.

TABLE OF DIPLOPIA

Name of

diplopia

Image of

right eye as

compared

with that of

the left is

CAUSED BY

CORRECTED BY

(1) By a natural

deviation of

(2) Artificially

by a prism

placed, base

(1) Turning

(2) Prism placed

with base

Lateral

Homonymous

On the

right

Either eye inward

(esophoria,

esotropia.)

In before either

eye.

Both eyes outward

(divergence.)

Out before either

eye.

Heteronymous

(or crossed)

On the

left

Either eye outward

(exophoria,

exotropia.)

Out before either

eye.

Both eyes inward

(convergence.)

In before either

eye.

Vertical

Right

Below

Right eye up

or left eye down

(right hyperphoria,

right hypertropia,

left hypotropia.)

Up before right

eye, down before

left eye.

Right eye down

and left eye up

(left supravergence.)

Down before right

eye or up before left eye.

Left

Above

Right eye up

or left eye down

(right hyperphoria,

right hypertropia,

left hypotropia.)

Up before right

eye, down before

left eye.

Right eye up and

left eye down

(rightsupravergence.)

Up before right

eye or down

before left.

MOVEMENT OF EACH EYE SINGLY

The movements of each eye individually are effected as follows:

The external rectus moves the eye directly outward; the internal rectus, directly inward.

The primary action of the superior rectus is to raise the eye. Because of the way in which the muscles run, obliquely from within outward, its lifting action increases when the eye is abducted and diminishes to little or nothing when the eye is adducted.

The inferior rectus carries the eye down. Owing to the oblique direction of the muscle, its depressing action increases as the eye is abducted and decreases to zero as the eye is adducted.

The inferior oblique is inserted back of the equator of the eye. Hence it pulls the back part of the eye down and consequently throws the front part up. It is thus an elevator of the eye, reinforcing the action of the superior rectus. Owing to the way in which it runs, from the front backward and outward, its elevating action is greatest when the eye is adducted, and diminishes to little or nothing when the eye is abducted.

The superior oblique, so far as its action on the eyeball is concerned, may be regarded as arising from the trochlea. From this point it runs backward and outward and is inserted back of the equator of the eye. It there pulls up the back part of the eye and consequently throws the front part down. It is thus a depressor, reinforcing the action of the inferior rectus. Owing to the oblique way in which it runs, its depressing action is greatest when the eye is adducted, and diminishes to little or nothing when the eye is abducted.

SUBSIDIARY ACTIONS

Besides these actions, rightly regarded as the main action of the ocular muscles, there are various subsidiary actions, due to the oblique way in which the superior and inferior recti and the two obliques run. Thus, both the superior and inferior recti adduct the eye, their action being most pronounced when the eye is already adducted. The two obliques, on the other hand, abduct the eye and do so most effectively when the eye is already abducted.

The superior rectus and superior oblique rotate the top of the vertical meridian of the eye inward (intorsion); while the inferior oblique and inferior rectus rotate it outward (extorsion). The superior and inferior recti act thus on the vertical meridian mainly when the eye is adducted; the oblique, on the other hand, when the eye is abducted.

Hence the eye is adducted by the internal rectus, assisted toward the end of its course by the superior and inferior recti. It is abducted by the external rectus, assisted toward the end of its course by the two obliques. It is carried straight up by the superior rectus and inferior oblique, up and out by the superior rectus and external rectus (the inferior oblique helping to carry it out, but not up; and in, mainly by the inferior oblique and internal rectus). The superior rectus assists in carrying it in, but hardly up at all.

The eye is likewise carried straight down by the inferior rectus and the superior oblique; down and out by the inferior and external recti, and down and in by the superior oblique and internal recti.

FIELD OF ACTION OF MUSCLES

As will be seen, each muscle acts most energetically in some special direction of the gaze, termed field of action of that particular muscle; thus the external rectus acts most powerfully when the eye is directed outward, and acts little or not at all when the eye is directed inward, except by purely passive traction. Likewise the superior rectus acts mainly when the eye is directed down. Furthermore, its action is limited to the upper and outer field; for in the upper and inner field elevation is performed chiefly by the inferior oblique.

This is also true of all the other muscles.

DIRECTION OF THE GAZE

There are six cardinal directions of the gaze, each corresponding to the field of action of one of the six ocular muscles as follows:

CARDINAL DIRECTION:

MUSCLES SPECIALLY ACTIVE:

Straight out

External rectus

Straight in

Internal rectus

Up and out

Superior rectus (as an elevator)

Up and in

Inferior oblique (as an elevator)

Down and out

Inferior rectus (as a depressor)

Down and in

Superior oblique (as a depressor)

It is to be noted that the action of each muscle does not absolutely stop at the middle line, but extends somewhat beyond it. Thus the action of the right externus extends not only throughout the whole right half of the field of vision, but also some fifteen to twenty degrees to the left of the median line; and that of the superior rectus extends not only above the horizontal plane but also somewhat below.

PRIMARY POSITION—FIELD OF FIXATION

Under normal conditions, when the head is erect and the eye is directed straight forward—that is, when its line of sight is perpendicular to the line joining the centres of rotation of the two eyes in the horizontal plane—the muscles are all balanced. This is called “the position of equilibrium” or the primary position. It is this position which must be assumed by the patient in conducting tests for balance of the muscles.

From the primary position, the eye may make excursions in every direction so that the patient can look at a whole series of objects in succession without moving the head. This portion of space, occupied by all the objects that may thus be seen directly by moving the eye without moving the head, is called “the field of fixation.”

BINOCULAR MOVEMENTS

While either eye alone may move in all possible directions, one cannot move independently of the other eye. Under ordinary circumstances, those movements only are possible which are regularly required to subserve binocular vision, hence, binocular single vision, as well. These movements are as follows:

PARALLEL MOVEMENTS

When one eye looks at a distant object the other is also directed to it, so that the lines of sight of the two eyes are parallel; if the distant object is moved about, the lines remain parallel, one moving as fast and as far as the other. These parallel movements of the two eyes are executed with considerable freedom in all directions, either eye being able to move readily to the right, left, up, down, or obliquely, provided the other eye moves precisely with it.

In executing any parallel movement, each eye is acted upon by at least three and sometimes by as many as five muscles. At times, but one of these muscles is required to produce any great movement of the eye, the others simply serving to steady it in its course. Thus when we look up to the right, although there are five muscles really acting upon each eye, the right eye is moved mainly by the external rectus and the left eye by the internal rectus.

Similarly, when we look up and to the right, although other muscles take part, the superior rectus is the chief muscle that moves the right eye up, and the external rectus the chief one that moves it to the right; while for the left eye the inferior oblique and the internal rectus are the efficient muscles.

A careful study of the action of the individual muscles will make it clear that these facts hold good for each of the cardinal directions of the gaze.

Furthermore, if we attentively consider the action of the twelve muscles moving the two eyes, we see that they may be divided into three groups, viz.; four lateral rotators, four elevators and four depressors.

LATERAL ROTATORS

Right rotators

to

Left rotators

L. Internal rectus

R. Internal rectus

R. External rectus

L. External rectus

ELEVATORS

Right-handed elevators

to

Left-handed elevators

(acting mainly when the

(acting mainly when the

eyes are directed to the right)

eyes are directed to the left)

R. Superior rectus

R. Inferior oblique

L. Inferior oblique

L. Superior rectus

DEPRESSORS

Right-handed depressors

to

Left-handed depressors

(acting mainly when the

(acting mainly when the

eyes are directed to the right)

eyes are directed to the left)

R. Inferior oblique

R. Superior oblique

L. Superior oblique

L. Inferior rectus.

Each group, it will be seen, comprises two pairs of muscles; one pair acting solely when the eyes are directed to the right, the other when they are directed to the left. It will further be noted that of the two muscles constituting any one pair, one is situated in the right eye, the other in the left.

EYE ASSOCIATES

The muscles forming any one pair are called associates. Any two associates acting together will move their respective eyes in precisely the same direction and to the same extent. Thus the right superior rectus moves the eye up to the left and rotates its vertical meridian to the left; and its associate, the left inferior oblique, moves its eye up to the left and rotates its vertical meridian to the left. This likewise applies to each of the other five groups of associates.

If one eye fails to keep pace with the other in executing parallel movements, diplopia ensues. If the eyes are moved in all directions and the point noted where the patient just begins to see double, we delimit the field of binocular single vision.

Normally, however, the two eyes maintain parallelism to the very limit of their excursion, so that diplopia occurs only at the extreme periphery of the field of vision, if at all. In fact, the field of binocular single vision usually extends not less than 40 degrees from the primary position in every direction.

Each of the various parallel movements of the eye appear to be governed by a distinct nerve mechanism, there being one centre for movements to the right, one for movements to the left, one for movements up, etc.

MOVEMENTS OF CONVERGENCE

In order to see an object at a nearby point, the eyes have to converge—a movement affected by a simultaneous and equal contraction of both internal recti. This movement may be combined with a vertical, lateral or oblique parallel movement. Thus, when we wish to look at a near object situated twenty degrees to our right, we first turn both eyes twenty degrees to the right, then converge both equally, turning the left a little more to the right and the right a little back toward the left.

Convergence is governed by a distinct mechanism of the nerves, the source of which has not been determined.

MOVEMENTS OF DIVERGENCE

In passing from a position of convergence to a position of parallelism, the lines of sight separate or diverge. This movement of divergence is a simultaneous, equal contraction of both externi; or, probably, of both actions combined. The eyes may even diverge somewhat beyond parallelism, as in overcoming prisms, base in, when looking at a distant object.

VERTICAL DIVERGENCE

The amount by which the lines of sight can separate in a vertical direction is very limited—at most but one or two degrees.

ORTHOPHORIA

The term orthophoria is used to denote an absolutely normal balance of the extrinsic muscles, just as the term emmetropia denotes a normal refractive condition. They are equally rare.

HETEROPHORIA

The term heterophoria includes all those conditions in which there is a tendency to depart from normal balance, but which nature is able to compensate for; while the term also includes the conditions in which nature has been unequal to the task and an actual turning or squint has occurred.

SUBDIVISIONS

The subdivisions of these terms at first reading appear complicated, but prove simple enough on closer study, indicating only the direction of the turning or tendency to turn. For instance:

- Esophoria signifies inward tendency.

- Exophoria signifies outward tendency.

- Hyperphoria signifies upward tendency.

- Hypophoria signifies downward tendency.

- Cyclophoria signifies tendency to torsion.

- Esotropia signifies inward turning.

- Exotropia signifies outward turning.

- Hypertropia signifies upward turning.

- Hypotropia signifies downward turning.

- Cyclotropia signifies actual torsion.

Combinations are describable in similar terms. A tendency of the right eye to turn up and inward, is a “right hyperesophoria”; the left eye to turn down and out, a “left hyperexophoria,” etc. Tendencies of both eyes together are denoted by the terms which follow:

- Anaphoria signifies an upward tendency.

- Kataphoria signifies a downward tendency.

- Dextrophoria signifies a right tendency.

- Laevophoria signifies a left tendency.

CHAPTER XVII

SYMPTOMS OF HETEROPHORIA

THESE depend on the kind of error present as well as the degree and widely vary.

In general, they may be said to fall into three classes—(1) defective vision, (2) pain of greater or less degree—(3) reflex symptoms.

Defective Vision. The first class may be present, even though each eye has a normal visual acuity; since, even when compensation is very good, the brain gets the impression of two objects, nearly, though not quite fused; and vision may be considerably worse with both eyes together than with either eye singly.

When compensation is considerably impaired, the diplopia becomes more and more persistent, till the brain finally makes choice of one image as more satisfactory, entirely suppressing the other. Visual acuity may not suffer in either eye; but vision being no longer binocular, everything is seen in the flat, the judgments of depth and distance being regularly more or less defective. While this is a tremendous disadvantage in many occupations, people gradually and not infrequently become accustomed to these visual defects and are not conscious of the handicap.

Pain. It is quite different with the second set of symptoms, which are always accompanied with pain. In fact, the character of the subjective symptoms in refractive errors and muscular imbalance is so similar that it is practically impossible to differentiate in many cases.

In muscular asthenopia, however, in addition to becoming easily tired, the patient often complains that letters seem to jump or run together or he may contend that he sees double for an instant; or again that he can “feel his eyes turn” involuntarily in their sockets. These pains or conditions are sometimes present only during actual use of the eyes. At other times they persist for hours. In some cases, after days or weeks of overstimulation, an explosion in migraine form occurs at irregular intervals. This condition often lasts a day or two.

Reflex Symptoms. In the third and last case, there are other reflex symptoms—such as dizziness, nausea, fainting, indigestion, insomnia and pains in other portions of the body—sometimes stimulating organic diseases.

The possibility of heterophoria as a factor in chorea, migraine, neurasthenia and other diseases which may be primarily due to unstable nerves, equilibrium is not to be forgotten. It is a notable fact that when the fusion compensation fails so completely that one image is entirely suppressed, or the diplopia is so great as to be overlooked, the symptoms often cease entirely.

TREATMENT

The treatment of heterophoria depends on a careful study of each individual case, but it cannot be too strongly emphasized that in the great majority of cases the subjective symptoms disappear after a full correction of the refraction is made.

In many cases, if the visual acuity in each eye be made normal, the fusion impulse alone will be sufficient to restore compensation.

Many cases of esophoria result from overstimulation of the centers for convergence and accommodation, made necessary by hyperopia and astigmatism, entirely disappearing when glasses abolish the need of accommodation. Cases of exophoria are sometimes due to the abnormal relaxation of accommodation and convergence which secures the best distant vision in myopia. Likewise the correction of myopia, by increasing the far point, may diminish the amount of convergence necessary for near vision.

Prisms for constant use are often prescribed, so placed as to help the weak muscles and counteract the strong. For instance, in esophoria we find the prism which, base in, will produce orthophoria for distance and prescribe a quarter of it, base in, before each eye. While this is very successful in some cases, the tendency in others is for the externus to increase slightly from constant exercise in overcoming the prism, while the internus decreases in proportion to the amount of work of which it is relieved. Prisms for permanent use are very beneficial in vertical deviations, since when the images are brought on the same level they require much less effort to secure fusion; and when prescribed base up or down, the effect secured is commonly an unchanging one.

We sometimes take advantage of this tendency when we prescribe for constant use weak prisms with the apex over the weak muscle, which gradually becomes strong from the exercise of overcoming it. This plan is effective only in patients who have a strong fusion impulse, and the prism selected must be weak enough to be easily overcome. We can accomplish the same effect by decentering the patient’s refraction lenses.

For instance, a convex lens so placed that the visual line passes the reverse will be the case if the lens is concave. The amount of prismatic action depends on the strength of the lens and the amount of decentering, the rule being that every centimeter of displacement causes as many prism diopters as there are diopters in that meridian of the lens. Thus +1 sphere, or cylinder axis 90, decentered one centimeter outward, is equivalent to adding a one degree prism diopter lens, base out.

DESTROPHORIA AND LAEVOPHORIA

These are terms denoting a condition in which both eyes are capable of abnormal rotating toward the right or left, as the case may be. The movement in the opposite direction is most common. The patient can often rotate his eyes 60 degrees toward the right, and to perhaps only 40 degrees to the left. His position of rest is parallel with his visual lines, but to the right, in looking at objects directly in front, he is much more comfortable with his head turned slightly to the left.

It is difficult to account for, except on the theory that definite movement of the eyes is rather to the right than to the left in most occupations. The position of the paper in writing at a desk tends toward dextrophoria; in reading, we move our eyes steadily from left to right and then begin a new line by a single brief movement to the left; the things that a man uses most—whether he be laborer or student—are kept within reach of the right hand, and in referring to them the eyes are constantly turned toward the right.

However, when these conditions result from other imbalances, they must be treated more carefully. For instance, a patient whose right internus is paralysed or congenitally defective on looking to the left, has a cross diplopia which vanishes to the right; as a result, he soon assumes a habit of carrying his head in this position. Ordinarily, this will cause no discomfort; but if the left internus is so weak that it cannot follow the right externus to its position of greatest ease, the visual lines are evidently different and the case must be treated as an exophoria.

If, on the other hand, the left internus over-balances the right externus, the condition is an esophoria and must be treated as such.

Similar reasoning applies to the conditions known as Anaphoria and Kataphoria, in which the visual lines are parallel to each other but directed up or down with regard to the horizontal plane of the body.

In the first, owing to congenital abnormalities, the eyes usually tend upward and the individual must go about with his chin on his chest, so that his eyes may look in front and yet remain in the position of rest. In the second, the chin is held in the air and the body arched backward.

But, unless extreme, neither of these conditions causes more than cosmetic difficulty and both should be undisturbed owing to the extreme difficulty of securing the same operative effect on both eyes. Suitable prisms are much more likely to be beneficial.

Supports for Holding

The Ski-optometer

Floor Stand

Wall Bracket

Chair Clamp

Chair Attachment

with Upright

Choice may be made from any of the above. The Wall Bracket is recommended, unless refractionist is provided with a specialist’s chair, to which the Chair Attachment with Upright may be attached.

Transcriber’s Notes:

The illustrations have been moved so that they do not break up paragraphs and so that they are next to the text they illustrate.

Typographical and punctuation errors have been silently corrected.

*** END OF THE PROJECT GUTENBERG EBOOK REFRACTION AND MUSCULAR IMBALANCE, AS SIMPLIFIED THROUGH THE USE OF THE SKI-OPTOMETER ***

Updated editions will replace the previous one—the old editions will be renamed.

Creating the works from print editions not protected by U.S. copyright law means that no one owns a United States copyright in these works, so the Foundation (and you!) can copy and distribute it in the United States without permission and without paying copyright royalties. Special rules, set forth in the General Terms of Use part of this license, apply to copying and distributing Project Gutenberg™ electronic works to protect the PROJECT GUTENBERG™ concept and trademark. Project Gutenberg is a registered trademark, and may not be used if you charge for an eBook, except by following the terms of the trademark license, including paying royalties for use of the Project Gutenberg trademark. If you do not charge anything for copies of this eBook, complying with the trademark license is very easy. You may use this eBook for nearly any purpose such as creation of derivative works, reports, performances and research. Project Gutenberg eBooks may be modified and printed and given away—you may do practically ANYTHING in the United States with eBooks not protected by U.S. copyright law. Redistribution is subject to the trademark license, especially commercial redistribution.

START: FULL LICENSE

THE FULL PROJECT GUTENBERG™ LICENSE

PLEASE READ THIS BEFORE YOU DISTRIBUTE OR USE THIS WORK

To protect the Project Gutenberg™ mission of promoting the free distribution of electronic works, by using or distributing this work (or any other work associated in any way with the phrase “Project Gutenberg”), you agree to comply with all the terms of the Full Project Gutenberg License available with this file or online at www.gutenberg.org/license.

Section 1. General Terms of Use and Redistributing Project Gutenberg electronic works

1.A. By reading or using any part of this Project Gutenberg electronic work, you indicate that you have read, understand, agree to and accept all the terms of this license and intellectual property (trademark/copyright) agreement. If you do not agree to abide by all the terms of this agreement, you must cease using and return or destroy all copies of Project Gutenberg electronic works in your possession. If you paid a fee for obtaining a copy of or access to a Project Gutenberg electronic work and you do not agree to be bound by the terms of this agreement, you may obtain a refund from the person or entity to whom you paid the fee as set forth in paragraph 1.E.8.

1.B. “Project Gutenberg” is a registered trademark. It may only be used on or associated in any way with an electronic work by people who agree to be bound by the terms of this agreement. There are a few things that you can do with most Project Gutenberg electronic works even without complying with the full terms of this agreement. See paragraph 1.C below. There are a lot of things you can do with Project Gutenberg electronic works if you follow the terms of this agreement and help preserve free future access to Project Gutenberg electronic works. See paragraph 1.E below.

1.C. The Project Gutenberg Literary Archive Foundation (“the Foundation” or PGLAF), owns a compilation copyright in the collection of Project Gutenberg electronic works. Nearly all the individual works in the collection are in the public domain in the United States. If an individual work is unprotected by copyright law in the United States and you are located in the United States, we do not claim a right to prevent you from copying, distributing, performing, displaying or creating derivative works based on the work as long as all references to Project Gutenberg are removed. Of course, we hope that you will support the Project Gutenberg mission of promoting free access to electronic works by freely sharing Project Gutenberg works in compliance with the terms of this agreement for keeping the Project Gutenberg name associated with the work. You can easily comply with the terms of this agreement by keeping this work in the same format with its attached full Project Gutenberg License when you share it without charge with others.

1.D. The copyright laws of the place where you are located also govern what you can do with this work. Copyright laws in most countries are in a constant state of change. If you are outside the United States, check the laws of your country in addition to the terms of this agreement before downloading, copying, displaying, performing, distributing or creating derivative works based on this work or any other Project Gutenberg work. The Foundation makes no representations concerning the copyright status of any work in any country other than the United States.

1.E. Unless you have removed all references to Project Gutenberg:

1.E.1. The following sentence, with active links to, or other immediate access to, the full Project Gutenberg License must appear prominently whenever any copy of a Project Gutenberg work (any work on which the phrase “Project Gutenberg” appears, or with which the phrase “Project Gutenberg” is associated) is accessed, displayed, performed, viewed, copied or distributed:

This eBook is for the use of anyone anywhere in the United States and most other parts of the world at no cost and with almost no restrictions whatsoever. You may copy it, give it away or re-use it under the terms of the Project Gutenberg™ License included with this eBook or online at www.gutenberg.org. If you are not located in the United States, you will have to check the laws of the country where you are located before using this eBook.

1.E.2. If an individual Project Gutenberg electronic work is derived from texts not protected by U.S. copyright law (does not contain a notice indicating that it is posted with permission of the copyright holder), the work can be copied and distributed to anyone in the United States without paying any fees or charges. If you are redistributing or providing access to a work with the phrase “Project Gutenberg” associated with or appearing on the work, you must comply either with the requirements of paragraphs 1.E.1 through 1.E.7 or obtain permission for the use of the work and the Project Gutenberg trademark as set forth in paragraphs 1.E.8 or 1.E.9.

1.E.3. If an individual Project Gutenberg electronic work is posted with the permission of the copyright holder, your use and distribution must comply with both paragraphs 1.E.1 through 1.E.7 and any additional terms imposed by the copyright holder. Additional terms will be linked to the Project Gutenberg License for all works posted with the permission of the copyright holder found at the beginning of this work.

1.E.4. Do not unlink or detach or remove the full Project Gutenberg License terms from this work, or any files containing a part of this work or any other work associated with Project Gutenberg.

1.E.5. Do not copy, display, perform, distribute or redistribute this electronic work, or any part of this electronic work, without prominently displaying the sentence set forth in paragraph 1.E.1 with active links or immediate access to the full terms of the Project Gutenberg License.

1.E.6. You may convert to and distribute this work in any binary, compressed, marked up, nonproprietary or proprietary form, including any word processing or hypertext form. However, if you provide access to or distribute copies of a Project Gutenberg work in a format other than “Plain Vanilla ASCII” or other format used in the official version posted on the official Project Gutenberg website (www.gutenberg.org), you must, at no additional cost, fee or expense to the user, provide a copy, a means of exporting a copy, or a means of obtaining a copy upon request, of the work in its original “Plain Vanilla ASCII” or other form. Any alternate format must include the full Project Gutenberg License as specified in paragraph 1.E.1.

1.E.7. Do not charge a fee for access to, viewing, displaying, performing, copying or distributing any Project Gutenberg works unless you comply with paragraph 1.E.8 or 1.E.9.

1.E.8. You may charge a reasonable fee for copies of or providing access to or distributing Project Gutenberg electronic works provided that:

- • You pay a royalty fee of 20% of the gross profits you derive from the use of Project Gutenberg works calculated using the method you already use to calculate your applicable taxes. The fee is owed to the owner of the Project Gutenberg trademark, but he has agreed to donate royalties under this paragraph to the Project Gutenberg Literary Archive Foundation. Royalty payments must be paid within 60 days following each date on which you prepare (or are legally required to prepare) your periodic tax returns. Royalty payments should be clearly marked as such and sent to the Project Gutenberg Literary Archive Foundation at the address specified in Section 4, “Information about donations to the Project Gutenberg Literary Archive Foundation.”

- • You provide a full refund of any money paid by a user who notifies you in writing (or by e-mail) within 30 days of receipt that s/he does not agree to the terms of the full Project Gutenberg™ License. You must require such a user to return or destroy all copies of the works possessed in a physical medium and discontinue all use of and all access to other copies of Project Gutenberg™ works.

- • You provide, in accordance with paragraph 1.F.3, a full refund of any money paid for a work or a replacement copy, if a defect in the electronic work is discovered and reported to you within 90 days of receipt of the work.

- • You comply with all other terms of this agreement for free distribution of Project Gutenberg™ works.

1.E.9. If you wish to charge a fee or distribute a Project Gutenberg™ electronic work or group of works on different terms than are set forth in this agreement, you must obtain permission in writing from the Project Gutenberg Literary Archive Foundation, the manager of the Project Gutenberg™ trademark. Contact the Foundation as set forth in Section 3 below.

1.F.

1.F.1. Project Gutenberg volunteers and employees expend considerable effort to identify, do copyright research on, transcribe and proofread works not protected by U.S. copyright law in creating the Project Gutenberg™ collection. Despite these efforts, Project Gutenberg™ electronic works, and the medium on which they may be stored, may contain “Defects,” such as, but not limited to, incomplete, inaccurate or corrupt data, transcription errors, a copyright or other intellectual property infringement, a defective or damaged disk or other medium, a computer virus, or computer codes that damage or cannot be read by your equipment.

1.F.2. LIMITED WARRANTY, DISCLAIMER OF DAMAGES - Except for the “Right of Replacement or Refund” described in paragraph 1.F.3, the Project Gutenberg Literary Archive Foundation, the owner of the Project Gutenberg™ trademark, and any other party distributing a Project Gutenberg™ electronic work under this agreement, disclaim all liability to you for damages, costs and expenses, including legal fees. YOU AGREE THAT YOU HAVE NO REMEDIES FOR NEGLIGENCE, STRICT LIABILITY, BREACH OF WARRANTY OR BREACH OF CONTRACT EXCEPT THOSE PROVIDED IN PARAGRAPH 1.F.3. YOU AGREE THAT THE FOUNDATION, THE TRADEMARK OWNER, AND ANY DISTRIBUTOR UNDER THIS AGREEMENT WILL NOT BE LIABLE TO YOU FOR ACTUAL, DIRECT, INDIRECT, CONSEQUENTIAL, PUNITIVE OR INCIDENTAL DAMAGES EVEN IF YOU GIVE NOTICE OF THE POSSIBILITY OF SUCH DAMAGE.

1.F.3. LIMITED RIGHT OF REPLACEMENT OR REFUND - If you discover a defect in this electronic work within 90 days of receiving it, you can receive a refund of the money (if any) you paid for it by sending a written explanation to the person you received the work from. If you received the work on a physical medium, you must return the medium with your written explanation. The person or entity that provided you with the defective work may elect to provide a replacement copy in lieu of a refund. If you received the work electronically, the person or entity providing it to you may choose to give you a second opportunity to receive the work electronically in lieu of a refund. If the second copy is also defective, you may demand a refund in writing without further opportunities to fix the problem.

1.F.4. Except for the limited right of replacement or refund set forth in paragraph 1.F.3, this work is provided to you ‘AS-IS’, WITH NO OTHER WARRANTIES OF ANY KIND, EXPRESS OR IMPLIED, INCLUDING BUT NOT LIMITED TO WARRANTIES OF MERCHANTABILITY OR FITNESS FOR ANY PURPOSE.

1.F.5. Some states do not allow disclaimers of certain implied warranties or the exclusion or limitation of certain types of damages. If any disclaimer or limitation set forth in this agreement violates the law of the state applicable to this agreement, the agreement shall be interpreted to make the maximum disclaimer or limitation permitted by the applicable state law. The invalidity or unenforceability of any provision of this agreement shall not void the remaining provisions.

1.F.6. INDEMNITY - You agree to indemnify and hold the Foundation, the trademark owner, any agent or employee of the Foundation, anyone providing copies of Project Gutenberg™ electronic works in accordance with this agreement, and any volunteers associated with the production, promotion and distribution of Project Gutenberg™ electronic works, harmless from all liability, costs and expenses, including legal fees, that arise directly or indirectly from any of the following which you do or cause to occur: (a) distribution of this or any Project Gutenberg work, (b) alteration, modification, or additions or deletions to any Project Gutenberg work, and (c) any Defect you cause.

Section 2. Information about the Mission of Project Gutenberg

Project Gutenberg is synonymous with the free distribution of electronic works in formats readable by the widest variety of computers including obsolete, old, middle-aged and new computers. It exists because of the efforts of hundreds of volunteers and donations from people in all walks of life.

Volunteers and financial support to provide volunteers with the assistance they need are critical to reaching Project Gutenberg’s goals and ensuring that the Project Gutenberg collection will remain freely available for generations to come. In 2001, the Project Gutenberg Literary Archive Foundation was created to provide a secure and permanent future for Project Gutenberg and future generations. To learn more about the Project Gutenberg Literary Archive Foundation and how your efforts and donations can help, see Sections 3 and 4 and the Foundation information page at www.gutenberg.org.

Section 3. Information about the Project Gutenberg Literary Archive Foundation

The Project Gutenberg Literary Archive Foundation is a non-profit 501(c)(3) educational corporation organized under the laws of the state of Mississippi and granted tax exempt status by the Internal Revenue Service. The Foundation’s EIN or federal tax identification number is 64-6221541. Contributions to the Project Gutenberg Literary Archive Foundation are tax deductible to the full extent permitted by U.S. federal laws and your state’s laws.

The Foundation’s business office is located at 41 Watchung Plaza #516, Montclair NJ 07042, USA, +1 (862) 621-9288. Email contact links and up to date contact information can be found at the Foundation’s website and official page at www.gutenberg.org/contact

Section 4. Information about Donations to the Project Gutenberg Literary Archive Foundation

Project Gutenberg™ depends upon and cannot survive without widespread public support and donations to carry out its mission of increasing the number of public domain and licensed works that can be freely distributed in machine-readable form accessible by the widest array of equipment including outdated equipment. Many small donations ($1 to $5,000) are particularly important to maintaining tax exempt status with the IRS.

The Foundation is committed to complying with the laws regulating charities and charitable donations in all 50 states of the United States. Compliance requirements are not uniform and it takes a considerable effort, much paperwork and many fees to meet and keep up with these requirements. We do not solicit donations in locations where we have not received written confirmation of compliance. To SEND DONATIONS or determine the status of compliance for any particular state visit www.gutenberg.org/donate.

While we cannot and do not solicit contributions from states where we have not met the solicitation requirements, we know of no prohibition against accepting unsolicited donations from donors in such states who approach us with offers to donate.

International donations are gratefully accepted, but we cannot make any statements concerning tax treatment of donations received from outside the United States. U.S. laws alone swamp our small staff.

Please check the Project Gutenberg web pages for current donation methods and addresses. Donations are accepted in a number of other ways including checks, online payments and credit card donations. To donate, please visit: www.gutenberg.org/donate.

Section 5. General Information About Project Gutenberg electronic works

Professor Michael S. Hart was the originator of the Project Gutenberg concept of a library of electronic works that could be freely shared with anyone. For forty years, he produced and distributed Project Gutenberg eBooks with only a loose network of volunteer support.

Project Gutenberg eBooks are often created from several printed editions, all of which are confirmed as not protected by copyright in the U.S. unless a copyright notice is included. Thus, we do not necessarily keep eBooks in compliance with any particular paper edition.

Most people start at our website which has the main PG search facility: www.gutenberg.org.

This website includes information about Project Gutenberg, including how to make donations to the Project Gutenberg Literary Archive Foundation, how to help produce our new eBooks, and how to subscribe to our email newsletter to hear about new eBooks.

Table of Contents

Refraction and Muscular Imbalance

INTRODUCTORY

Chapter I SKI-OPTOMETER CONSTRUCTION

Convex Spherical Lenses

Operates and Indicates Automatically

Concave Spherical Lenses

Chapter II CYLINDRICAL LENSES

Obtaining Correct Focus

Why Concave Cylinders Are Used Exclusively

Transposition of Lenses

Chapter III HOW THE SKI-OPTOMETER ASSISTS IN REFRACTION

The Use of the Ski-optometer in Skioscopy

A Simplified Skioscopic Method

Employing Spheres and Cylinders in Skioscopy

Use of the Ski-optometer in Subjective Testing

A Simplified Subjective Method

Procedure for Using Minus Cylinders Exclusively

Constant Attention Not Required

Chapter IV IMPORTANT POINTS IN CONNECTION WITH THE USE OF THE SKI-OPTOMETER

Elimination of Trial-Frame Discomfort

Rigidity of Construction

How to Place the Ski-optometer in Position

Cleaning the Lenses

Accuracy Assured in Every Test

Built to Last a Lifetime

Chapter V CONDENSED PROCEDURE FOR MAKING SPHERE AND CYLINDER TEST WITH THE SKI-OPTOMETER

Subjective Distance Test

Subjective Reading Test

Chapter VI MUSCULAR IMBALANCE

The Action of Prisms

The Phorometer

The Maddox Rod

Procedure for Making the Muscle Test

Binocular and Monocular Test

Chapter VII THE BINOCULAR MUSCLE TEST

Made with the Maddox Rod and Phorometer

Esophoria and Exophoria

Making Muscle Test Before and After Optical Correction

When to Consider Correction of Muscular Imbalance

Four Methods for Correction of Muscular Imbalance

The Rotary Prism

Use of the Rotary Prism in Binocular Muscle Tests

Chapter VIII THE MONOCULAR DUCTION MUSCLE TEST

Made with Both Rotary Prisms

Locating the Faulty Muscle

Adduction

Abduction

Superduction

Subduction

Procedure for Monocular Muscle Testing

Diagnosing a Specific Muscle Case

Chapter IX FIRST METHOD OF TREATMENT OPTICAL CORRECTION

Esophoria

Treatment for Correcting Esophoria in Children

Chapter X SECOND METHOD OF TREATMENT— MUSCULAR EXERCISE

Exophoria

An Assumed Case

Effect of Muscular Exercise

Home Treatment for Muscular Exercise— Square Prism Set Used in Conjunction With the Ski-Optometer

Chapter XI THIRD METHOD OF TREATMENT—PRISM LENSES

When and How Employed

Prism Reduction Method

Chapter XII A CONDENSATION OF PREVIOUS CHAPTERS ON THE PROCEDURE FOR MUSCLE TESTING WITH THE SKI-OPTOMETER

Four Methods of Treating an Imbalance Case When the Preceding One Fails

Prisms

Cyclophoria

Chapter XIII CYCLOPHORIA

Made with Maddox Rods and Rotary Prisms

Chapter XIV CYCLODUCTION TEST

Made With the Combined Use of the Two Maddox Rods

Treatment for Cyclophoria

Chapter XV MOVEMENTS OF THE EYEBALLS AND THEIR ANOMALIES

Monocular Fixation

Binocular Fixation

Orthophoria

Heterophoria

Squint

Varieties of Heterophoria and Squint

Chapter XVI LAW OF PROJECTION

Suppression of Image

Monocular Diplopia

Movement of Each Eye Singly

Subsidiary Actions

Field of Action of Muscles

Direction of the Gaze

Primary Position—Field of Fixation

Binocular Movements

Parallel Movements

Eye Associates

Movements of Convergence

Movements of Divergence

Vertical Divergence

Orthophoria

Heterophoria

Subdivisions

Chapter XVII SYMPTOMS OF HETEROPHORIA

Treatment

Destrophoria and Laevophoria

THE FULL PROJECT GUTENBERG™ LICENSE